-- ERROR --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 16:54:52 2024 UTC
-- File 20240207_120440_1229.dat, 25510  packets found.
APID:
      1229 (EW) in 25510 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 25510 packets (all packets)
Total number of packets: 
      36720 in 25510 packets (all packets)
Packet No from 1 to 25510
!!!ERROR: Maximum packet number must be = 36720
   Sequence of packet numbers -- OK -- 
!!!ERROR: Last packet number must be = 36720
Hz and MHz counters from 19.257472 to 77.953572
-each separately from 19s to 77s and from 46118us to 953572us.
   Differences in neigboring packets: 
      0.000000 in 25506 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -41.907454 in 1 packets: 2448
      -16.788646 in 1 packets: 7548
      16.788646 in 1 packets: 19788
MIU packet time from 2024-02-07T12:05:02.257.472.000 to 2024-02-07T12:06:00.953.572.000
   Differences in neigboring packets: 
      0.000000 in 25506 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -41.907454 in 1 packets: 2448
      -16.788646 in 1 packets: 7548
      16.788646 in 1 packets: 19788
MIU orbit number: 
      60000 in 25510 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 25510 packets (all packets)
MIU orbit information / number of events: 
      0 in 25510 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 25510 packets (all packets)
Instrument anomalies: 
      0 (OK) in 25510 packets (all packets)
FPGA configuration version: 
      211 in 25510 packets (all packets)
10 MHz counter at the last 1Hz pulse from 20.8750065 to 78.8748071
   Differences in neigboring packets: 
      0.0000000 in 25506 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -40.9998588 in 1 packets: 2448
      -16.9999418 in 1 packets: 7548
      16.9999418 in 1 packets: 19788
Temperature FPGA: 
      05 -> 48.5 degC in 13270 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 12240 packets: 7549, 7550, 7551, 7552, 7553, 7554, 7555, 7556, 7557, 7558, 7559, 7560, 7561, 7562, 7563, 7564, 7565, 7566, ...
Temperature XO: 
      C0 -> 35.6 degC in 25510 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 25510 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 25510 packets (all packets)
Number of external events: 
      3 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 10822 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
      1 in 12240 packets: 7549, 7550, 7551, 7552, 7553, 7554, 7555, 7556, 7557, 7558, 7559, 7560, 7561, 7562, 7563, 7564, 7565, 7566, ...
Number of generated alerts: 
      3 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 10822 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
      1 in 12240 packets: 7549, 7550, 7551, 7552, 7553, 7554, 7555, 7556, 7557, 7558, 7559, 7560, 7561, 7562, 7563, 7564, 7565, 7566, ...
Survey memory error: 
      0 in 25510 packets (all packets)
Event memory error: 
      0 in 25510 packets (all packets)
Configuration error: 
      0 in 25510 packets (all packets)
SDRAM CRC error: 
      0 in 25510 packets (all packets)
Radio gain: 
      0 in 25510 packets (all packets)
Radio bandwidth: 
      0 in 25510 packets (all packets)
Digital gain: 
      0 in 25510 packets (all packets)
Analog gain: 
      0 in 25510 packets (all packets)

EVENT DATA READER
-- File 20240207_120440_1229.dat, 125 EW pages found. Extra packets: 10.
   ERROR -  Extra packets
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      3 in 2448 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 10812 packets: 2449, 2450, 2451, 2452, 2453, 2454, 2455, 2456, 2457, 2458, 2459, 2460, 2461, 2462, 2463, 2464, 2465, 2466, ...
      1 in 12240 packets: 7549, 7550, 7551, 7552, 7553, 7554, 7555, 7556, 7557, 7558, 7559, 7560, 7561, 7562, 7563, 7564, 7565, 7566, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 21.0769911 to 79.8144901
      Difference between the neighboring packets: 
      0.0000170 in 25496 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      -41.9489110 in 1 packets: 2448
      -16.8752710 in 1 packets: 7548
      16.6672250 in 1 packets: 19788
   Memory page number from 9 to 133
      Difference between the neighboring packets: 
      0 in 25375 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 124 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 25375 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 121 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2652, 2856, 3060, 3264, 3468, 3672, 3876, ...
      -11 in 1 packets: 2448
      -24 in 1 packets: 7548
      -34 in 1 packets: 19788
   MIU event flag 
      1 in 25500 packets (all packets)
   Time triggered event flag 
      0 in 25500 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2024-02-07T12:05:02.201.983.637
   Min EW header abs. time found among all events            : 2024-02-07T12:05:02.257.472.000
   Max EW event (1st sample) abs. time found among all events: 2024-02-07T12:06:00.898.083.637
   Max EW header abs. time found among all events            : 2024-02-07T12:06:00.953.572.000
   TM values from -248 to 207 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      65 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      66 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      67 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      68 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      69 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      70 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      71 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      72 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      73 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      74 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      75 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      76 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      77 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      78 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      79 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -248 to 194 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=19 C1MHz=0.257472
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 21.0769911
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-02-07T12:05:02.257.472.000
   Event 1st sample abs. time : 2024-02-07T12:05:02.201.983.637)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 19, 32bit 10MHz counter = 21.076991100 corrected by 40bit 10MHz counter of 20.875006500 obtained at 1Hz counter of 19. Result: 19.201985287
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              19.201985112
   Event data descriptor XO time (reconstructed seconds from XP start)    :              19.201985287
   Sampling frequency correction factor for this event:     0.999996600

 Event no. 2 total number of packets: 10812
    Page no. 1:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      125 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      126 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      127 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      128 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      129 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      130 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      131 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      132 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      133 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 53  ordered packets: 10812
   TM values from -247 to 207 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=36 C1MHz=0.046118
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 10811 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 37.8655791
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-02-07T12:05:19.046.118.000
   Event 1st sample abs. time : 2024-02-07T12:05:18.990.629.631)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 36, 32bit 10MHz counter = 37.865579100 corrected by 40bit 10MHz counter of 37.874948300 obtained at 1Hz counter of 36. Result: 35.990630767
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              35.990630592
   Event data descriptor XO time (reconstructed seconds from XP start)    :              35.990630767
   Sampling frequency correction factor for this event:     0.999996500

 Event no. 3 total number of packets: 2448
    Page no. 1:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 12  ordered packets: 2448
   TM values from -217 to 187 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=77 C1MHz=0.953572
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 2447 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 79.7728911
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2024-02-07T12:06:00.953.572.000
   Event 1st sample abs. time : 2024-02-07T12:06:00.898.083.637)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 77, 32bit 10MHz counter = 79.772891100 corrected by 40bit 10MHz counter of 74.874820800 obtained at 1Hz counter of 73. Result: 77.898086953
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              77.898086778
   Event data descriptor XO time (reconstructed seconds from XP start)    :              77.898086953
   Sampling frequency correction factor for this event:     0.999996600
--
Extremes in separate events:

 - Event number: 1 from 2024-02-07T12:05:02.201.983.637 to 2024-02-07T12:05:02.410.064.344 (XO: 19.201985112 - 19.410065819)
   Min TM value:-248 at an offset of 54799.361 usec -> 2024-02-07T12:05:02.256.782.998 (XO: 19.256784473)
   Max TM value:194 at an offset of 54799.049 usec -> 2024-02-07T12:05:02.256.782.685 (XO: 19.256784161)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 144818.167 usec -> 2024-02-07T12:05:02.346.801.804 (XO: 19.346803279)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 54799.199 usec -> 2024-02-07T12:05:02.256.782.835 (XO: 19.256784311)
   Med: -4.00000   Ave: -4.4342243   Std: 4.5151101

 - Event number: 2 from 2024-02-07T12:05:18.990.629.631 to 2024-02-07T12:05:19.174.434.274 (XO: 35.990630592 - 36.174435236)
   Min TM value:-247 at an offset of 84628.596 usec -> 2024-02-07T12:05:19.075.258.227 (XO: 36.075259188)
   Max TM value:207 at an offset of 84628.321 usec -> 2024-02-07T12:05:19.075.257.952 (XO: 36.075258913)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 54645.829 usec -> 2024-02-07T12:05:19.045.275.459 (XO: 36.045276421)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 84628.446 usec -> 2024-02-07T12:05:19.075.258.077 (XO: 36.075259038)
   Med: -4.00000   Ave: -4.4401532   Std: 4.0963339

 - Event number: 3 from 2024-02-07T12:06:00.898.083.637 to 2024-02-07T12:06:00.939.699.778 (XO: 77.898086778 - 77.939702920)
   Min TM value:-217 at an offset of 32946.362 usec -> 2024-02-07T12:06:00.931.029.999 (XO: 77.931033141)
   Max TM value:187 at an offset of 32946.062 usec -> 2024-02-07T12:06:00.931.029.699 (XO: 77.931032841)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 32943.787 usec -> 2024-02-07T12:06:00.931.027.424 (XO: 77.931030566)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 32946.175 usec -> 2024-02-07T12:06:00.931.029.811 (XO: 77.931032953)
   Med: -4.00000   Ave: -4.4333363   Std: 4.0070872
