-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 10:59:28 2024 UTC
-- File 20240206_142140_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.047375 to 53.235160
-each separately from 0s to 53s and from 47375us to 490204us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.442251 in 1 packets: 18
      13.771679 in 1 packets: 36
      21.972699 in 1 packets: 54
MIU packet time from 2024-02-06T14:21:43.047.375.000 to 2024-02-06T14:22:36.235.160.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.442251 in 1 packets: 18
      13.771679 in 1 packets: 36
      21.972699 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8198733 to 54.8196902
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999413 in 1 packets: 18
      13.9999516 in 1 packets: 36
      21.9999240 in 1 packets: 54
Temperature FPGA: 
      07 -> 45.9 degC in 72 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240206_142140_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8672498 to 1.8675388
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4421910  -- OK for SL --
   MIU time from 2024-02-06T14:21:43.047.375.000 to 2024-02-06T14:21:43.047.664.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.442251

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3097298 to 19.3100188
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7716310  -- OK for SL --
   MIU time from 2024-02-06T14:22:00.489.915.000 to 2024-02-06T14:22:00.490.204.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.771679

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.0816498 to 33.0819388
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.9726230  -- OK for SL --
   MIU time from 2024-02-06T14:22:14.261.883.000 to 2024-02-06T14:22:14.262.172.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.972699

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.0545618 to 55.0548508
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-06T14:22:36.234.871.000 to 2024-02-06T14:22:36.235.160.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.867249800 corrected by 40bit 10MHz counter of 1.819873300 obtained at 1Hz counter of 0. Result: 0.047376666
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-06T14:21:43.047.234.725
   Max 1st sample MIU time among all SL packets: 2024-02-06T14:22:36.235.019.725
   Min 1st sample XO time among all SL packets: 0.047236390
   Max 1st sample XO time among all SL packets: 53.235021124
   TM values from -267 to 238 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-06T14:21:43.047.234.725 (XO: 0.047236390 - 0.047542391, t=34.0 degC)  Rank: 00077
   TM values: Max =   235  Min =  -250  Ave =    -4.54  Std =    18.29
   Min TM value:-250 at an offset of 121.838 usec -> 2024-02-06T14:21:43.047.356.562 (XO: 0.047358228)
   Max TM value:235 at an offset of 121.575 usec -> 2024-02-06T14:21:43.047.356.300 (XO: 0.047357966)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 119.300 usec -> 2024-02-06T14:21:43.047.354.025 (XO: 0.047355691)
   Min dTM/dt [1/us]:-2.53e+03 at an offset of 121.688 usec -> 2024-02-06T14:21:43.047.356.412 (XO: 0.047358078)

 - Block number: 2 starting at 2024-02-06T14:22:00.489.774.725 (XO: 17.489776639 - 17.490082640, t=34.0 degC)  Rank: 00080
   TM values: Max =   216  Min =  -267  Ave =    -4.45  Std =    30.55
   Min TM value:-267 at an offset of 181.813 usec -> 2024-02-06T14:22:00.489.956.538 (XO: 17.489958452)
   Max TM value:216 at an offset of 108.200 usec -> 2024-02-06T14:22:00.489.882.925 (XO: 17.489884840)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 111.038 usec -> 2024-02-06T14:22:00.489.885.762 (XO: 17.489887677)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 181.663 usec -> 2024-02-06T14:22:00.489.956.388 (XO: 17.489958302)

 - Block number: 3 starting at 2024-02-06T14:22:14.261.742.725 (XO: 31.261744215 - 31.262050216, t=34.0 degC)  Rank: 00083
   TM values: Max =   203  Min =  -234  Ave =    -4.51  Std =    17.57
   Min TM value:-234 at an offset of 110.513 usec -> 2024-02-06T14:22:14.261.853.237 (XO: 31.261854728)
   Max TM value:203 at an offset of 107.788 usec -> 2024-02-06T14:22:14.261.850.512 (XO: 31.261852003)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 110.613 usec -> 2024-02-06T14:22:14.261.853.337 (XO: 31.261854828)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 107.900 usec -> 2024-02-06T14:22:14.261.850.625 (XO: 31.261852115)

 - Block number: 4 starting at 2024-02-06T14:22:36.234.730.725 (XO: 53.234732123 - 53.235038124, t=34.0 degC)  Rank: 00082
   TM values: Max =   238  Min =  -265  Ave =    -4.46  Std =    18.29
   Min TM value:-265 at an offset of 119.625 usec -> 2024-02-06T14:22:36.234.850.350 (XO: 53.234851748)
   Max TM value:238 at an offset of 119.338 usec -> 2024-02-06T14:22:36.234.850.062 (XO: 53.234851461)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 122.138 usec -> 2024-02-06T14:22:36.234.852.862 (XO: 53.234854261)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 119.463 usec -> 2024-02-06T14:22:36.234.850.187 (XO: 53.234851586)
