-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 06:23:40 2024 UTC
-- File 20240205_133045_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.732047 to 23.330815
-each separately from 0s to 23s and from 330526us to 732336us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.598190 in 1 packets: 18
MIU packet time from 2024-02-05T13:30:48.732.047.000 to 2024-02-05T13:31:11.330.815.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.598190 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8765541 to 24.8764747
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999206 in 1 packets: 18
Temperature FPGA: 
      09 -> 43.3 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      08 -> 44.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240205_133045_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.6086013 to 2.6088903
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5981110  -- OK for SL --
   MIU time from 2024-02-05T13:30:48.732.047.000 to 2024-02-05T13:30:48.732.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.598190

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2070013 to 25.2072903
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-05T13:31:11.330.526.000 to 2024-02-05T13:31:11.330.815.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.608601300 corrected by 40bit 10MHz counter of 1.876554100 obtained at 1Hz counter of 0. Result: 0.732049689
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-05T13:30:48.731.906.725
   Max 1st sample MIU time among all SL packets: 2024-02-05T13:31:11.330.674.725
   Min 1st sample XO time among all SL packets: 0.731909413
   Max 1st sample XO time among all SL packets: 23.330676482
   TM values from -265 to 259 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-05T13:30:48.731.906.725 (XO: 0.731909413 - 0.732215415, t=32.4 degC)  Rank: 00082
   TM values: Max =   258  Min =  -265  Ave =    -4.53  Std =    17.96
   Min TM value:-265 at an offset of 117.038 usec -> 2024-02-05T13:30:48.732.023.762 (XO: 0.732026451)
   Max TM value:258 at an offset of 116.763 usec -> 2024-02-05T13:30:48.732.023.487 (XO: 0.732026176)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 116.675 usec -> 2024-02-05T13:30:48.732.023.400 (XO: 0.732026089)
   Min dTM/dt [1/us]:-2.69e+03 at an offset of 116.863 usec -> 2024-02-05T13:30:48.732.023.587 (XO: 0.732026276)

 - Block number: 2 starting at 2024-02-05T13:31:11.330.385.725 (XO: 23.330387481 - 23.330693482, t=32.4 degC)  Rank: 00080
   TM values: Max =   259  Min =  -265  Ave =    -4.56  Std =    17.99
   Min TM value:-265 at an offset of 115.238 usec -> 2024-02-05T13:31:11.330.500.962 (XO: 23.330502719)
   Max TM value:259 at an offset of 114.988 usec -> 2024-02-05T13:31:11.330.500.712 (XO: 23.330502469)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 114.900 usec -> 2024-02-05T13:31:11.330.500.625 (XO: 23.330502382)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 115.100 usec -> 2024-02-05T13:31:11.330.500.825 (XO: 23.330502582)
