-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 06:13:11 2024 UTC
-- File 20240205_132116_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.033741 to 78.667516
-each separately from 0s to 78s and from 33741us to 961169us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.257928 in 1 packets: 18
      12.668633 in 1 packets: 36
      18.998225 in 1 packets: 54
      11.178615 in 1 packets: 72
      14.528640 in 1 packets: 90
MIU packet time from 2024-02-05T13:21:19.033.741.000 to 2024-02-05T13:22:37.667.516.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.257928 in 1 packets: 18
      12.668633 in 1 packets: 36
      18.998225 in 1 packets: 54
      11.178615 in 1 packets: 72
      14.528640 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9274267 to 79.9271579
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999276 in 1 packets: 18
      11.9999587 in 1 packets: 36
      18.9999345 in 1 packets: 54
      11.9999588 in 1 packets: 72
      13.9999516 in 1 packets: 90
Temperature FPGA: 
      08 -> 44.6 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      07 -> 45.9 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      BE -> 32.4 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 89 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20240205_132116_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9611689 to 1.9614579
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2578550  -- OK for SL --
   MIU time from 2024-02-05T13:21:19.033.741.000 to 2024-02-05T13:21:19.034.030.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.257928

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2193129 to 23.2196019
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6685910  -- OK for SL --
   MIU time from 2024-02-05T13:21:40.291.958.000 to 2024-02-05T13:21:40.292.247.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.668633

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.8881929 to 35.8884819
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9981590  -- OK for SL --
   MIU time from 2024-02-05T13:21:52.960.880.000 to 2024-02-05T13:21:52.961.169.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.998225

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.8866409 to 54.8869299
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1785750  -- OK for SL --
   MIU time from 2024-02-05T13:22:11.959.394.000 to 2024-02-05T13:22:11.959.683.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.178615

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.0655049 to 66.0657939
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5285910  -- OK for SL --
   MIU time from 2024-02-05T13:22:23.138.298.000 to 2024-02-05T13:22:23.138.587.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.528640

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.5943849 to 80.5946739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-05T13:22:37.667.227.000 to 2024-02-05T13:22:37.667.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.961168900 corrected by 40bit 10MHz counter of 1.927426700 obtained at 1Hz counter of 0. Result: 0.033742315
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-05T13:21:19.033.600.725
   Max 1st sample MIU time among all SL packets: 2024-02-05T13:22:37.667.375.725
   Min 1st sample XO time among all SL packets: 0.033602039
   Max 1st sample XO time among all SL packets: 78.667377994
   TM values from -267 to 251 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-05T13:21:19.033.600.725 (XO: 0.033602039 - 0.033908040, t=32.4 degC)  Rank: 00080
   TM values: Max =   202  Min =  -261  Ave =    -4.43  Std =    18.78
   Min TM value:-261 at an offset of 123.425 usec -> 2024-02-05T13:21:19.033.724.150 (XO: 0.033725465)
   Max TM value:202 at an offset of 111.050 usec -> 2024-02-05T13:21:19.033.711.775 (XO: 0.033713090)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 126.075 usec -> 2024-02-05T13:21:19.033.726.800 (XO: 0.033728115)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 111.175 usec -> 2024-02-05T13:21:19.033.711.900 (XO: 0.033713215)

 - Block number: 2 starting at 2024-02-05T13:21:40.291.817.725 (XO: 21.291819317 - 21.292125318, t=32.4 degC)  Rank: 00083
   TM values: Max =   216  Min =  -267  Ave =    -4.55  Std =    22.33
   Min TM value:-267 at an offset of 125.175 usec -> 2024-02-05T13:21:40.291.942.900 (XO: 21.291944493)
   Max TM value:216 at an offset of 124.888 usec -> 2024-02-05T13:21:40.291.942.612 (XO: 21.291944205)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 127.788 usec -> 2024-02-05T13:21:40.291.945.512 (XO: 21.291947105)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 125.013 usec -> 2024-02-05T13:21:40.291.942.737 (XO: 21.291944330)

 - Block number: 3 starting at 2024-02-05T13:21:52.960.739.725 (XO: 33.960742988 - 33.961048989, t=32.4 degC)  Rank: 00080
   TM values: Max =   209  Min =  -235  Ave =    -4.49  Std =    21.65
   Min TM value:-235 at an offset of 127.450 usec -> 2024-02-05T13:21:52.960.867.175 (XO: 33.960870438)
   Max TM value:209 at an offset of 124.675 usec -> 2024-02-05T13:21:52.960.864.400 (XO: 33.960867663)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 127.563 usec -> 2024-02-05T13:21:52.960.867.287 (XO: 33.960870551)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 124.800 usec -> 2024-02-05T13:21:52.960.864.525 (XO: 33.960867788)

 - Block number: 4 starting at 2024-02-05T13:22:11.959.253.725 (XO: 52.959256386 - 52.959562388, t=32.4 degC)  Rank: 00078
   TM values: Max =   233  Min =  -226  Ave =    -4.42  Std =    23.22
   Min TM value:-226 at an offset of 115.600 usec -> 2024-02-05T13:22:11.959.369.325 (XO: 52.959371987)
   Max TM value:233 at an offset of 112.875 usec -> 2024-02-05T13:22:11.959.366.600 (XO: 52.959369262)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 115.713 usec -> 2024-02-05T13:22:11.959.369.437 (XO: 52.959372099)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 112.975 usec -> 2024-02-05T13:22:11.959.366.700 (XO: 52.959369362)

 - Block number: 5 starting at 2024-02-05T13:22:23.138.157.725 (XO: 64.138158795 - 64.138464796, t=32.4 degC)  Rank: 00079
   TM values: Max =   214  Min =  -236  Ave =    -4.53  Std =    19.74
   Min TM value:-236 at an offset of 125.163 usec -> 2024-02-05T13:22:23.138.282.887 (XO: 64.138283958)
   Max TM value:214 at an offset of 122.425 usec -> 2024-02-05T13:22:23.138.280.150 (XO: 64.138281220)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 125.275 usec -> 2024-02-05T13:22:23.138.283.000 (XO: 64.138284070)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 122.550 usec -> 2024-02-05T13:22:23.138.280.275 (XO: 64.138281345)

 - Block number: 6 starting at 2024-02-05T13:22:37.667.086.725 (XO: 78.667088993 - 78.667394994, t=32.4 degC)  Rank: 00075
   TM values: Max =   251  Min =  -244  Ave =    -4.54  Std =    18.77
   Min TM value:-244 at an offset of 126.663 usec -> 2024-02-05T13:22:37.667.213.387 (XO: 78.667215656)
   Max TM value:251 at an offset of 126.388 usec -> 2024-02-05T13:22:37.667.213.112 (XO: 78.667215381)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 126.300 usec -> 2024-02-05T13:22:37.667.213.025 (XO: 78.667215294)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 126.488 usec -> 2024-02-05T13:22:37.667.213.212 (XO: 78.667215481)
