-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 05:43:01 2024 UTC
-- File 20240205_125159_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.882121 to 23.832748
-each separately from 13s to 23s and from 832459us to 882410us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.950049 in 1 packets: 18
MIU packet time from 2024-02-05T12:52:15.882.121.000 to 2024-02-05T12:52:25.832.748.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.950049 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8875289 to 24.8874944
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999655 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240205_125159_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.7696495 to 15.7699385
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9500150  -- OK for SL --
   MIU time from 2024-02-05T12:52:15.882.121.000 to 2024-02-05T12:52:15.882.410.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.950049

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.7199535 to 25.7202425
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-05T12:52:25.832.459.000 to 2024-02-05T12:52:25.832.748.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.769649500 corrected by 40bit 10MHz counter of 14.887528900 obtained at 1Hz counter of 13. Result: 13.882123599
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-05T12:52:15.881.980.725
   Max 1st sample MIU time among all SL packets: 2024-02-05T12:52:25.832.607.725
   Min 1st sample XO time among all SL packets: 13.881983324
   Max 1st sample XO time among all SL packets: 23.832610739
   TM values from -258 to 231 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-05T12:52:15.881.980.725 (XO: 13.881983324 - 13.882289325, t=32.4 degC)  Rank: 00084
   TM values: Max =   231  Min =  -258  Ave =    -4.40  Std =    18.48
   Min TM value:-258 at an offset of 129.025 usec -> 2024-02-05T12:52:15.882.109.750 (XO: 13.882112349)
   Max TM value:231 at an offset of 128.750 usec -> 2024-02-05T12:52:15.882.109.475 (XO: 13.882112074)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 131.625 usec -> 2024-02-05T12:52:15.882.112.350 (XO: 13.882114949)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 128.863 usec -> 2024-02-05T12:52:15.882.109.587 (XO: 13.882112187)

 - Block number: 2 starting at 2024-02-05T12:52:25.832.318.725 (XO: 23.832321738 - 23.832627739, t=32.4 degC)  Rank: 00078
   TM values: Max =   213  Min =  -228  Ave =    -4.49  Std =    23.06
   Min TM value:-228 at an offset of 129.775 usec -> 2024-02-05T12:52:25.832.448.500 (XO: 23.832451514)
   Max TM value:213 at an offset of 127.025 usec -> 2024-02-05T12:52:25.832.445.750 (XO: 23.832448764)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 129.875 usec -> 2024-02-05T12:52:25.832.448.600 (XO: 23.832451614)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 127.125 usec -> 2024-02-05T12:52:25.832.445.850 (XO: 23.832448864)
