-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 02:40:08 2024 UTC
-- File 20240204_144733_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.018441 to 20.879255
-each separately from 3s to 20s and from 18441us to 879255us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.860236 in 1 packets: 18
MIU packet time from 2024-02-04T14:47:39.018.441.000 to 2024-02-04T14:47:56.879.255.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.860236 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8284746 to 21.8284156
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999410 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      08 -> 44.6 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240204_144733_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8469171 to 4.8472061
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8601750  -- OK for SL --
   MIU time from 2024-02-04T14:47:39.018.441.000 to 2024-02-04T14:47:39.018.730.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.860236

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7073811 to 22.7076701
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-04T14:47:56.878.966.000 to 2024-02-04T14:47:56.879.255.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.846917100 corrected by 40bit 10MHz counter of 4.828474600 obtained at 1Hz counter of 3. Result: 3.018442563
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-04T14:47:39.018.300.725
   Max 1st sample MIU time among all SL packets: 2024-02-04T14:47:56.879.114.725
   Min 1st sample XO time among all SL packets: 3.018302287
   Max 1st sample XO time among all SL packets: 20.879117214
   TM values from -286 to 271 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-04T14:47:39.018.300.725 (XO: 3.018302287 - 3.018608288, t=34.0 degC)  Rank: 00083
   TM values: Max =   263  Min =  -286  Ave =    -4.44  Std =    19.77
   Min TM value:-286 at an offset of 122.888 usec -> 2024-02-04T14:47:39.018.423.612 (XO: 3.018425175)
   Max TM value:263 at an offset of 122.638 usec -> 2024-02-04T14:47:39.018.423.362 (XO: 3.018424925)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 125.450 usec -> 2024-02-04T14:47:39.018.426.175 (XO: 3.018427738)
   Min dTM/dt [1/us]:-2.79e+03 at an offset of 122.738 usec -> 2024-02-04T14:47:39.018.423.462 (XO: 3.018425025)

 - Block number: 2 starting at 2024-02-04T14:47:56.878.825.725 (XO: 20.878828213 - 20.879134214, t=34.0 degC)  Rank: 00081
   TM values: Max =   271  Min =  -284  Ave =    -4.46  Std =    21.13
   Min TM value:-284 at an offset of 115.913 usec -> 2024-02-04T14:47:56.878.941.637 (XO: 20.878944126)
   Max TM value:271 at an offset of 115.663 usec -> 2024-02-04T14:47:56.878.941.387 (XO: 20.878943876)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 115.563 usec -> 2024-02-04T14:47:56.878.941.287 (XO: 20.878943776)
   Min dTM/dt [1/us]:-2.80e+03 at an offset of 115.763 usec -> 2024-02-04T14:47:56.878.941.487 (XO: 20.878943976)
