-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 02:34:55 2024 UTC
-- File 20240204_144250_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.448535 to 20.438657
-each separately from 13s to 20s and from 438368us to 448824us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.989544 in 1 packets: 18
MIU packet time from 2024-02-04T14:43:06.448.535.000 to 2024-02-04T14:43:13.438.657.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.989544 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8443120 to 21.8442877
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999757 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      06 -> 47.2 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240204_144250_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.2928480 to 15.2931370
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9895190  -- OK for SL --
   MIU time from 2024-02-04T14:43:06.448.535.000 to 2024-02-04T14:43:06.448.824.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.989544

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.2826560 to 22.2829450
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-04T14:43:13.438.368.000 to 2024-02-04T14:43:13.438.657.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.292848000 corrected by 40bit 10MHz counter of 14.844312000 obtained at 1Hz counter of 13. Result: 13.448537570
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-02-04T14:43:06.448.394.725
   Max 1st sample MIU time among all SL packets: 2024-02-04T14:43:13.438.516.725
   Min 1st sample XO time among all SL packets: 13.448397294
   Max 1st sample XO time among all SL packets: 20.438518560
   TM values from -266 to 234 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-04T14:43:06.448.394.725 (XO: 13.448397294 - 13.448703295, t=34.0 degC)  Rank: 00089
   TM values: Max =   207  Min =  -266  Ave =    -4.48  Std =    17.47
   Min TM value:-266 at an offset of 109.688 usec -> 2024-02-04T14:43:06.448.504.412 (XO: 13.448506982)
   Max TM value:207 at an offset of 106.938 usec -> 2024-02-04T14:43:06.448.501.662 (XO: 13.448504232)
   Max dTM/dt [1/us]:2.58e+03 at an offset of 109.788 usec -> 2024-02-04T14:43:06.448.504.512 (XO: 13.448507082)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 107.038 usec -> 2024-02-04T14:43:06.448.501.762 (XO: 13.448504332)

 - Block number: 2 starting at 2024-02-04T14:43:13.438.227.725 (XO: 20.438229559 - 20.438535560, t=34.0 degC)  Rank: 00080
   TM values: Max =   234  Min =  -265  Ave =    -4.47  Std =    19.57
   Min TM value:-265 at an offset of 121.225 usec -> 2024-02-04T14:43:13.438.348.950 (XO: 20.438350784)
   Max TM value:234 at an offset of 108.938 usec -> 2024-02-04T14:43:13.438.336.662 (XO: 20.438338497)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 123.888 usec -> 2024-02-04T14:43:13.438.351.612 (XO: 20.438353447)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 109.038 usec -> 2024-02-04T14:43:13.438.336.762 (XO: 20.438338597)
