-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb 11 00:03:50 2024 UTC
-- File 20240204_090206_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.225972 to 15.575142
-each separately from 1s to 15s and from 225972us to 575142us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.348592 in 1 packets: 18
MIU packet time from 2024-02-04T09:02:10.225.972.000 to 2024-02-04T09:02:24.575.142.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.348592 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4814948 to 16.4814463
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999515 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240204_090206_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7074674 to 2.7077564
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3485430  -- OK for SL --
   MIU time from 2024-02-04T09:02:10.225.972.000 to 2024-02-04T09:02:10.226.261.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.348592

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0562994 to 17.0565884
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-04T09:02:24.574.853.000 to 2024-02-04T09:02:24.575.142.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.707467400 corrected by 40bit 10MHz counter of 2.481494800 obtained at 1Hz counter of 1. Result: 1.225973391
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-02-04T09:02:10.225.831.725
   Max 1st sample MIU time among all SL packets: 2024-02-04T09:02:24.575.001.725
   Min 1st sample XO time among all SL packets: 1.225833115
   Max 1st sample XO time among all SL packets: 15.575003838
   TM values from -232 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-04T09:02:10.225.831.725 (XO: 1.225833115 - 1.226139116, t=34.0 degC)  Rank: 00075
   TM values: Max =   177  Min =  -208  Ave =    -4.48  Std =    18.25
   Min TM value:-208 at an offset of 118.763 usec -> 2024-02-04T09:02:10.225.950.487 (XO: 1.225951878)
   Max TM value:177 at an offset of 118.413 usec -> 2024-02-04T09:02:10.225.950.137 (XO: 1.225951528)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 121.188 usec -> 2024-02-04T09:02:10.225.952.912 (XO: 1.225954303)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 118.525 usec -> 2024-02-04T09:02:10.225.950.250 (XO: 1.225951641)

 - Block number: 2 starting at 2024-02-04T09:02:24.574.712.725 (XO: 15.574714837 - 15.575020838, t=34.0 degC)  Rank: 00074
   TM values: Max =   193  Min =  -232  Ave =    -4.48  Std =    16.38
   Min TM value:-232 at an offset of 127.650 usec -> 2024-02-04T09:02:24.574.840.375 (XO: 15.574842487)
   Max TM value:193 at an offset of 127.338 usec -> 2024-02-04T09:02:24.574.840.062 (XO: 15.574842174)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 130.125 usec -> 2024-02-04T09:02:24.574.842.850 (XO: 15.574844962)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 127.463 usec -> 2024-02-04T09:02:24.574.840.187 (XO: 15.574842299)
