-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Feb  6 19:13:41 2024 UTC
-- File 20240203_120644_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.889014 to 21.248182
-each separately from 7s to 21s and from 247893us to 889303us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.358590 in 1 packets: 18
MIU packet time from 2024-02-03T12:06:54.889.014.000 to 2024-02-03T12:07:08.248.182.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.358590 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.9272439 to 22.9271958
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999519 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 36 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240203_120644_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8162575 to 9.8165465
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3585430  -- OK for SL --
   MIU time from 2024-02-03T12:06:54.889.014.000 to 2024-02-03T12:06:54.889.303.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.358590

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1750895 to 23.1753785
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-03T12:07:08.247.893.000 to 2024-02-03T12:07:08.248.182.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.816257500 corrected by 40bit 10MHz counter of 8.927243900 obtained at 1Hz counter of 7. Result: 7.889016623
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-02-03T12:06:54.888.873.725
   Max 1st sample MIU time among all SL packets: 2024-02-03T12:07:08.248.041.725
   Min 1st sample XO time among all SL packets: 7.888876347
   Max 1st sample XO time among all SL packets: 21.248043293
   TM values from -239 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-03T12:06:54.888.873.725 (XO: 7.888876347 - 7.889182348, t=34.0 degC)  Rank: 00079
   TM values: Max =   196  Min =  -228  Ave =    -4.58  Std =    18.03
   Min TM value:-228 at an offset of 115.325 usec -> 2024-02-03T12:06:54.888.989.050 (XO: 7.888991673)
   Max TM value:196 at an offset of 115.000 usec -> 2024-02-03T12:06:54.888.988.725 (XO: 7.888991348)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 117.850 usec -> 2024-02-03T12:06:54.888.991.575 (XO: 7.888994198)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 115.100 usec -> 2024-02-03T12:06:54.888.988.825 (XO: 7.888991448)

 - Block number: 2 starting at 2024-02-03T12:07:08.247.752.725 (XO: 21.247754292 - 21.248060293, t=34.0 degC)  Rank: 00084
   TM values: Max =   204  Min =  -239  Ave =    -4.58  Std =    15.06
   Min TM value:-239 at an offset of 128.063 usec -> 2024-02-03T12:07:08.247.880.787 (XO: 21.247882355)
   Max TM value:204 at an offset of 127.738 usec -> 2024-02-03T12:07:08.247.880.462 (XO: 21.247882030)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 130.525 usec -> 2024-02-03T12:07:08.247.883.250 (XO: 21.247884818)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 127.888 usec -> 2024-02-03T12:07:08.247.880.612 (XO: 21.247882180)
