-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Feb  6 14:14:04 2024 UTC
-- File 20240201_140601_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.349143 to 39.130542
-each separately from 0s to 39s and from 130253us to 429834us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.080113 in 1 packets: 18
      19.700419 in 1 packets: 36
MIU packet time from 2024-02-01T14:06:04.349.143.000 to 2024-02-01T14:06:43.130.542.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.080113 in 1 packets: 18
      19.700419 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7828056 to 40.7826707
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999344 in 1 packets: 18
      19.9999307 in 1 packets: 36
Temperature FPGA: 
      08 -> 44.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      BF -> 34.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240201_140601_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1319491 to 2.1322381
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0800470  -- OK for SL --
   MIU time from 2024-02-01T14:06:04.349.143.000 to 2024-02-01T14:06:04.349.432.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.080113

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.2122851 to 21.2125741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7003510  -- OK for SL --
   MIU time from 2024-02-01T14:06:23.429.545.000 to 2024-02-01T14:06:23.429.834.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.700419

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.9129251 to 40.9132141
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-02-01T14:06:43.130.253.000 to 2024-02-01T14:06:43.130.542.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.131949100 corrected by 40bit 10MHz counter of 1.782805600 obtained at 1Hz counter of 0. Result: 0.349144722
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-02-01T14:06:04.349.002.725
   Max 1st sample MIU time among all SL packets: 2024-02-01T14:06:43.130.401.725
   Min 1st sample XO time among all SL packets: 0.349004447
   Max 1st sample XO time among all SL packets: 39.130403581
   TM values from -256 to 225 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-02-01T14:06:04.349.002.725 (XO: 0.349004447 - 0.349310448, t=34.0 degC)  Rank: 00080
   TM values: Max =   214  Min =  -255  Ave =    -4.65  Std =    21.87
   Min TM value:-255 at an offset of 126.663 usec -> 2024-02-01T14:06:04.349.129.387 (XO: 0.349131109)
   Max TM value:214 at an offset of 126.363 usec -> 2024-02-01T14:06:04.349.129.087 (XO: 0.349130809)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 129.213 usec -> 2024-02-01T14:06:04.349.131.937 (XO: 0.349133659)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 126.488 usec -> 2024-02-01T14:06:04.349.129.212 (XO: 0.349130934)

 - Block number: 2 starting at 2024-02-01T14:06:23.429.404.725 (XO: 19.429406328 - 19.429712329, t=34.0 degC)  Rank: 00081
   TM values: Max =   225  Min =  -256  Ave =    -4.60  Std =    19.03
   Min TM value:-256 at an offset of 128.650 usec -> 2024-02-01T14:06:23.429.533.375 (XO: 19.429534978)
   Max TM value:225 at an offset of 128.388 usec -> 2024-02-01T14:06:23.429.533.112 (XO: 19.429534716)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 126.138 usec -> 2024-02-01T14:06:23.429.530.862 (XO: 19.429532466)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 128.488 usec -> 2024-02-01T14:06:23.429.533.212 (XO: 19.429534816)

 - Block number: 3 starting at 2024-02-01T14:06:43.130.112.725 (XO: 39.130114580 - 39.130420581, t=34.0 degC)  Rank: 00082
   TM values: Max =   220  Min =  -251  Ave =    -4.52  Std =    18.59
   Min TM value:-251 at an offset of 129.063 usec -> 2024-02-01T14:06:43.130.241.787 (XO: 39.130243643)
   Max TM value:220 at an offset of 128.763 usec -> 2024-02-01T14:06:43.130.241.487 (XO: 39.130243343)
   Max dTM/dt [1/us]:1.45e+03 at an offset of 126.513 usec -> 2024-02-01T14:06:43.130.239.237 (XO: 39.130241093)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 128.863 usec -> 2024-02-01T14:06:43.130.241.587 (XO: 39.130243443)
