-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 13:47:46 2024 UTC
-- File 20240122_125224_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.812782 to 25.259511
-each separately from 4s to 25s and from 259222us to 813071us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.446151 in 1 packets: 18
MIU packet time from 2024-01-22T12:52:31.812.782.000 to 2024-01-22T12:52:52.259.511.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.446151 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8629583 to 26.8628860
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999277 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 36 packets (all packets)
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240122_125224_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.6757404 to 6.6760294
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.4460790  -- OK for SL --
   MIU time from 2024-01-22T12:52:31.812.782.000 to 2024-01-22T12:52:31.813.071.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.446151

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.1221084 to 27.1223974
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-22T12:52:52.259.222.000 to 2024-01-22T12:52:52.259.511.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.675740400 corrected by 40bit 10MHz counter of 5.862958300 obtained at 1Hz counter of 4. Result: 4.812784945
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-01-22T12:52:31.812.641.725
   Max 1st sample MIU time among all SL packets: 2024-01-22T12:52:52.259.370.725
   Min 1st sample XO time among all SL packets: 4.812644669
   Max 1st sample XO time among all SL packets: 25.259372033
   TM values from -256 to 258 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-22T12:52:31.812.641.725 (XO: 4.812644669 - 4.812950670, t=32.4 degC)  Rank: 00085
   TM values: Max =   206  Min =  -254  Ave =    -4.54  Std =    20.72
   Min TM value:-254 at an offset of 113.675 usec -> 2024-01-22T12:52:31.812.755.400 (XO: 4.812758345)
   Max TM value:206 at an offset of 110.900 usec -> 2024-01-22T12:52:31.812.752.625 (XO: 4.812755570)
   Max dTM/dt [1/us]:2.52e+03 at an offset of 113.775 usec -> 2024-01-22T12:52:31.812.755.500 (XO: 4.812758445)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 111.038 usec -> 2024-01-22T12:52:31.812.752.762 (XO: 4.812755707)

 - Block number: 2 starting at 2024-01-22T12:52:52.259.081.725 (XO: 25.259083032 - 25.259389033, t=32.4 degC)  Rank: 00081
   TM values: Max =   258  Min =  -256  Ave =    -4.62  Std =    19.73
   Min TM value:-256 at an offset of 126.450 usec -> 2024-01-22T12:52:52.259.208.175 (XO: 25.259209482)
   Max TM value:258 at an offset of 126.213 usec -> 2024-01-22T12:52:52.259.207.937 (XO: 25.259209245)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 129.088 usec -> 2024-01-22T12:52:52.259.210.812 (XO: 25.259212120)
   Min dTM/dt [1/us]:-2.70e+03 at an offset of 126.313 usec -> 2024-01-22T12:52:52.259.208.037 (XO: 25.259209345)
