-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 13:41:22 2024 UTC
-- File 20240122_124510_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.576057 to 40.641490
-each separately from 10s to 40s and from 153647us to 641490us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.577301 in 1 packets: 18
      14.487265 in 1 packets: 36
MIU packet time from 2024-01-22T12:45:24.576.057.000 to 2024-01-22T12:45:54.641.490.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.577301 in 1 packets: 18
      14.487265 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.3033462 to 42.3032424
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999447 in 1 packets: 18
      13.9999515 in 1 packets: 36
Temperature FPGA: 
      08 -> 44.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      09 -> 43.3 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      BE -> 32.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240122_124510_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.8794030 to 12.8796920
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5772470  -- OK for SL --
   MIU time from 2024-01-22T12:45:24.576.057.000 to 2024-01-22T12:45:24.576.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.577301

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.4569390 to 28.4572280
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4872150  -- OK for SL --
   MIU time from 2024-01-22T12:45:40.153.647.000 to 2024-01-22T12:45:40.153.936.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.487265

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.9444430 to 42.9447320
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-22T12:45:54.641.201.000 to 2024-01-22T12:45:54.641.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.879403000 corrected by 40bit 10MHz counter of 12.303346200 obtained at 1Hz counter of 10. Result: 10.576058816
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-22T12:45:24.575.916.725
   Max 1st sample MIU time among all SL packets: 2024-01-22T12:45:54.641.349.725
   Min 1st sample XO time among all SL packets: 10.575918541
   Max 1st sample XO time among all SL packets: 40.641351506
   TM values from -253 to 262 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-22T12:45:24.575.916.725 (XO: 10.575918541 - 10.576224542, t=32.4 degC)  Rank: 00078
   TM values: Max =   243  Min =  -221  Ave =    -4.59  Std =    19.18
   Min TM value:-221 at an offset of 129.500 usec -> 2024-01-22T12:45:24.576.046.225 (XO: 10.576048041)
   Max TM value:243 at an offset of 126.750 usec -> 2024-01-22T12:45:24.576.043.475 (XO: 10.576045291)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 129.613 usec -> 2024-01-22T12:45:24.576.046.337 (XO: 10.576048154)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 126.850 usec -> 2024-01-22T12:45:24.576.043.575 (XO: 10.576045391)

 - Block number: 2 starting at 2024-01-22T12:45:40.153.506.725 (XO: 26.153508362 - 26.153814363, t=32.4 degC)  Rank: 00081
   TM values: Max =   262  Min =  -245  Ave =    -4.54  Std =    24.81
   Min TM value:-245 at an offset of 125.638 usec -> 2024-01-22T12:45:40.153.632.362 (XO: 26.153634000)
   Max TM value:262 at an offset of 127.988 usec -> 2024-01-22T12:45:40.153.634.712 (XO: 26.153636350)
   Max dTM/dt [1/us]:2.72e+03 at an offset of 125.750 usec -> 2024-01-22T12:45:40.153.632.475 (XO: 26.153634113)
   Min dTM/dt [1/us]:-2.65e+03 at an offset of 128.075 usec -> 2024-01-22T12:45:40.153.634.800 (XO: 26.153636438)

 - Block number: 3 starting at 2024-01-22T12:45:54.641.060.725 (XO: 40.641062505 - 40.641368506, t=32.4 degC)  Rank: 00080
   TM values: Max =   259  Min =  -253  Ave =    -4.62  Std =    19.74
   Min TM value:-253 at an offset of 129.538 usec -> 2024-01-22T12:45:54.641.190.262 (XO: 40.641192043)
   Max TM value:259 at an offset of 129.275 usec -> 2024-01-22T12:45:54.641.190.000 (XO: 40.641191780)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 129.188 usec -> 2024-01-22T12:45:54.641.189.912 (XO: 40.641191693)
   Min dTM/dt [1/us]:-2.69e+03 at an offset of 129.375 usec -> 2024-01-22T12:45:54.641.190.100 (XO: 40.641191880)
