-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 11:56:11 2024 UTC
-- File 20240122_100210_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.890421 to 24.091763
-each separately from 6s to 24s and from 91474us to 890710us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.200764 in 1 packets: 18
MIU packet time from 2024-01-22T10:02:19.890.421.000 to 2024-01-22T10:02:37.091.763.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.200764 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8614927 to 25.8614305
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999378 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      09 -> 43.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240122_100210_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7519137 to 8.7522027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2007030  -- OK for SL --
   MIU time from 2024-01-22T10:02:19.890.421.000 to 2024-01-22T10:02:19.890.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.200764

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.9529057 to 25.9531947
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-22T10:02:37.091.474.000 to 2024-01-22T10:02:37.091.763.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.751913700 corrected by 40bit 10MHz counter of 7.861492700 obtained at 1Hz counter of 6. Result: 6.890424116
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-22T10:02:19.890.280.725
   Max 1st sample MIU time among all SL packets: 2024-01-22T10:02:37.091.622.725
   Min 1st sample XO time among all SL packets: 6.890283841
   Max 1st sample XO time among all SL packets: 24.091624237
   TM values from -253 to 247 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-22T10:02:19.890.280.725 (XO: 6.890283841 - 6.890589842, t=32.4 degC)  Rank: 00083
   TM values: Max =   247  Min =  -253  Ave =    -4.54  Std =    19.11
   Min TM value:-253 at an offset of 119.925 usec -> 2024-01-22T10:02:19.890.400.650 (XO: 6.890403766)
   Max TM value:247 at an offset of 117.188 usec -> 2024-01-22T10:02:19.890.397.912 (XO: 6.890401029)
   Max dTM/dt [1/us]:2.50e+03 at an offset of 120.038 usec -> 2024-01-22T10:02:19.890.400.762 (XO: 6.890403879)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 117.288 usec -> 2024-01-22T10:02:19.890.398.012 (XO: 6.890401129)

 - Block number: 2 starting at 2024-01-22T10:02:37.091.333.725 (XO: 24.091335236 - 24.091641237, t=32.4 degC)  Rank: 00080
   TM values: Max =   228  Min =  -247  Ave =    -4.61  Std =    28.65
   Min TM value:-247 at an offset of 121.450 usec -> 2024-01-22T10:02:37.091.455.175 (XO: 24.091456686)
   Max TM value:228 at an offset of 92.700 usec -> 2024-01-22T10:02:37.091.426.425 (XO: 24.091427936)
   Max dTM/dt [1/us]:2.36e+03 at an offset of 121.550 usec -> 2024-01-22T10:02:37.091.455.275 (XO: 24.091456786)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 92.800 usec -> 2024-01-22T10:02:37.091.426.525 (XO: 24.091428036)
