-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 09:26:20 2024 UTC
-- File 20240122_054930_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.681396 to 24.753247
-each separately from 12s to 24s and from 681396us to 753247us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.071273 in 1 packets: 18
MIU packet time from 2024-01-22T05:49:45.681.396.000 to 2024-01-22T05:49:57.753.247.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.071273 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8513780 to 25.8513365
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999585 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      BF -> 34.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240122_054930_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.5327746 to 14.5330636
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0712310  -- OK for SL --
   MIU time from 2024-01-22T05:49:45.681.396.000 to 2024-01-22T05:49:45.681.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.071273

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.6042946 to 26.6045836
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-22T05:49:57.752.958.000 to 2024-01-22T05:49:57.753.247.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.532774600 corrected by 40bit 10MHz counter of 13.851378000 obtained at 1Hz counter of 12. Result: 12.681398985
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-22T05:49:45.681.255.725
   Max 1st sample MIU time among all SL packets: 2024-01-22T05:49:57.753.106.725
   Min 1st sample XO time among all SL packets: 12.681258709
   Max 1st sample XO time among all SL packets: 24.753109386
   TM values from -261 to 273 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-22T05:49:45.681.255.725 (XO: 12.681258709 - 12.681564710, t=32.4 degC)  Rank: 00082
   TM values: Max =   209  Min =  -247  Ave =    -4.60  Std =    21.13
   Min TM value:-247 at an offset of 129.088 usec -> 2024-01-22T05:49:45.681.384.812 (XO: 12.681387797)
   Max TM value:209 at an offset of 131.738 usec -> 2024-01-22T05:49:45.681.387.462 (XO: 12.681390447)
   Max dTM/dt [1/us]:2.59e+03 at an offset of 131.625 usec -> 2024-01-22T05:49:45.681.387.350 (XO: 12.681390335)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 128.938 usec -> 2024-01-22T05:49:45.681.384.662 (XO: 12.681387647)

 - Block number: 2 starting at 2024-01-22T05:49:57.752.817.725 (XO: 24.752820385 - 24.753126386, t=34.0 degC)  Rank: 00081
   TM values: Max =   273  Min =  -261  Ave =    -4.48  Std =    19.93
   Min TM value:-261 at an offset of 126.600 usec -> 2024-01-22T05:49:57.752.944.325 (XO: 24.752946985)
   Max TM value:273 at an offset of 126.363 usec -> 2024-01-22T05:49:57.752.944.087 (XO: 24.752946748)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 126.275 usec -> 2024-01-22T05:49:57.752.944.000 (XO: 24.752946660)
   Min dTM/dt [1/us]:-3.00e+03 at an offset of 126.475 usec -> 2024-01-22T05:49:57.752.944.200 (XO: 24.752946860)
