-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 07:20:32 2024 UTC
-- File 20240122_021042_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.220047 to 52.420927
-each separately from 6s to 52s and from 220047us to 730668us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.349923 in 1 packets: 18
      16.159831 in 1 packets: 36
      14.689970 in 1 packets: 54
MIU packet time from 2024-01-22T02:10:51.220.047.000 to 2024-01-22T02:11:37.420.927.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.349923 in 1 packets: 18
      16.159831 in 1 packets: 36
      14.689970 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8904962 to 53.8903372
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999481 in 1 packets: 18
      15.9999446 in 1 packets: 36
      14.9999483 in 1 packets: 54
Temperature FPGA: 
      07 -> 45.9 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      08 -> 44.6 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      BF -> 34.0 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20240122_021042_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.1105437 to 8.1108327
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3498710  -- OK for SL --
   MIU time from 2024-01-22T02:10:51.220.047.000 to 2024-01-22T02:10:51.220.336.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.349923

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.4607037 to 23.4609927
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1597750  -- OK for SL --
   MIU time from 2024-01-22T02:11:06.570.259.000 to 2024-01-22T02:11:06.570.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.159831

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.6207677 to 39.6210567
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6899190  -- OK for SL --
   MIU time from 2024-01-22T02:11:22.730.379.000 to 2024-01-22T02:11:22.730.668.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.689970

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.3109757 to 54.3112647
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-22T02:11:37.420.638.000 to 2024-01-22T02:11:37.420.927.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.110543700 corrected by 40bit 10MHz counter of 7.890496200 obtained at 1Hz counter of 6. Result: 6.220048248
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-22T02:10:51.219.906.725
   Max 1st sample MIU time among all SL packets: 2024-01-22T02:11:37.420.786.725
   Min 1st sample XO time among all SL packets: 6.219907973
   Max 1st sample XO time among all SL packets: 52.420788698
   TM values from -268 to 265 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-22T02:10:51.219.906.725 (XO: 6.219907973 - 6.220213974, t=34.0 degC)  Rank: 00074
   TM values: Max =   223  Min =  -239  Ave =    -4.44  Std =    24.64
   Min TM value:-239 at an offset of 127.125 usec -> 2024-01-22T02:10:51.220.033.850 (XO: 6.220035098)
   Max TM value:223 at an offset of 112.338 usec -> 2024-01-22T02:10:51.220.019.062 (XO: 6.220020311)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 127.250 usec -> 2024-01-22T02:10:51.220.033.975 (XO: 6.220035223)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 112.463 usec -> 2024-01-22T02:10:51.220.019.187 (XO: 6.220020436)

 - Block number: 2 starting at 2024-01-22T02:11:06.570.118.725 (XO: 21.570121063 - 21.570427064, t=34.0 degC)  Rank: 00080
   TM values: Max =   265  Min =  -268  Ave =    -4.48  Std =    20.88
   Min TM value:-268 at an offset of 114.538 usec -> 2024-01-22T02:11:06.570.233.262 (XO: 21.570235601)
   Max TM value:265 at an offset of 114.300 usec -> 2024-01-22T02:11:06.570.233.025 (XO: 21.570235364)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 117.150 usec -> 2024-01-22T02:11:06.570.235.875 (XO: 21.570238214)
   Min dTM/dt [1/us]:-2.78e+03 at an offset of 114.375 usec -> 2024-01-22T02:11:06.570.233.100 (XO: 21.570235439)

 - Block number: 3 starting at 2024-01-22T02:11:22.730.238.725 (XO: 37.730241008 - 37.730547009, t=34.0 degC)  Rank: 00080
   TM values: Max =   255  Min =  -253  Ave =    -4.51  Std =    23.94
   Min TM value:-253 at an offset of 126.725 usec -> 2024-01-22T02:11:22.730.365.450 (XO: 37.730367733)
   Max TM value:255 at an offset of 126.463 usec -> 2024-01-22T02:11:22.730.365.187 (XO: 37.730367471)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 126.388 usec -> 2024-01-22T02:11:22.730.365.112 (XO: 37.730367396)
   Min dTM/dt [1/us]:-2.81e+03 at an offset of 126.575 usec -> 2024-01-22T02:11:22.730.365.300 (XO: 37.730367583)

 - Block number: 4 starting at 2024-01-22T02:11:37.420.497.725 (XO: 52.420499697 - 52.420805698, t=34.0 degC)  Rank: 00084
   TM values: Max =   211  Min =  -245  Ave =    -4.54  Std =    19.93
   Min TM value:-245 at an offset of 127.050 usec -> 2024-01-22T02:11:37.420.624.775 (XO: 52.420626747)
   Max TM value:211 at an offset of 111.650 usec -> 2024-01-22T02:11:37.420.609.375 (XO: 52.420611347)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 127.150 usec -> 2024-01-22T02:11:37.420.624.875 (XO: 52.420626847)
   Min dTM/dt [1/us]:-1.95e+03 at an offset of 124.375 usec -> 2024-01-22T02:11:37.420.622.100 (XO: 52.420624072)
