-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb  5 05:08:18 2024 UTC
-- File 20240121_213325_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.786838 to 23.143521
-each separately from 6s to 23s and from 143232us to 787127us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.356105 in 1 packets: 18
MIU packet time from 2024-01-21T21:33:34.786.838.000 to 2024-01-21T21:33:51.143.521.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.356105 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8416201 to 24.8415614
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999413 in 1 packets: 18
Temperature FPGA: 
      07 -> 45.9 degC in 36 packets (all packets)
Temperature XO: 
      BF -> 34.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240121_213325_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6284584 to 8.6287474
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3560470  -- OK for SL --
   MIU time from 2024-01-21T21:33:34.786.838.000 to 2024-01-21T21:33:34.787.127.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.356105

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9847944 to 24.9850834
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-21T21:33:51.143.232.000 to 2024-01-21T21:33:51.143.521.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.628458400 corrected by 40bit 10MHz counter of 7.841620100 obtained at 1Hz counter of 6. Result: 6.786841054
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2024-01-21T21:33:34.786.697.725
   Max 1st sample MIU time among all SL packets: 2024-01-21T21:33:51.143.380.725
   Min 1st sample XO time among all SL packets: 6.786700778
   Max 1st sample XO time among all SL packets: 23.143382227
   TM values from -283 to 238 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-21T21:33:34.786.697.725 (XO: 6.786700778 - 6.787006780, t=34.0 degC)  Rank: 00080
   TM values: Max =   238  Min =  -283  Ave =    -4.43  Std =    20.67
   Min TM value:-283 at an offset of 119.600 usec -> 2024-01-21T21:33:34.786.817.325 (XO: 6.786820379)
   Max TM value:238 at an offset of 107.325 usec -> 2024-01-21T21:33:34.786.805.050 (XO: 6.786808104)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 122.175 usec -> 2024-01-21T21:33:34.786.819.900 (XO: 6.786822954)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 107.438 usec -> 2024-01-21T21:33:34.786.805.162 (XO: 6.786808216)

 - Block number: 2 starting at 2024-01-21T21:33:51.143.091.725 (XO: 23.143093226 - 23.143399227, t=34.0 degC)  Rank: 00082
   TM values: Max =   236  Min =  -274  Ave =    -4.53  Std =    25.27
   Min TM value:-274 at an offset of 118.113 usec -> 2024-01-21T21:33:51.143.209.837 (XO: 23.143211339)
   Max TM value:236 at an offset of 117.813 usec -> 2024-01-21T21:33:51.143.209.537 (XO: 23.143211039)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 120.675 usec -> 2024-01-21T21:33:51.143.212.400 (XO: 23.143213901)
   Min dTM/dt [1/us]:-2.68e+03 at an offset of 117.950 usec -> 2024-01-21T21:33:51.143.209.675 (XO: 23.143211176)
