-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb  4 23:57:30 2024 UTC
-- File 20240121_082528_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.969183 to 40.438310
-each separately from 6s to 40s and from 438021us to 969472us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.549176 in 1 packets: 18
      21.919084 in 1 packets: 36
MIU packet time from 2024-01-21T08:25:37.969.183.000 to 2024-01-21T08:26:11.438.310.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.549176 in 1 packets: 18
      21.919084 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8152943 to 41.8151763
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999584 in 1 packets: 18
      21.9999236 in 1 packets: 36
Temperature FPGA: 
      08 -> 44.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      07 -> 45.9 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      BE -> 32.4 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      BF -> 34.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20240121_082528_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7844774 to 8.7847664
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.5491350  -- OK for SL --
   MIU time from 2024-01-21T08:25:37.969.183.000 to 2024-01-21T08:25:37.969.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.549176

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3339014 to 20.3341904
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.9190070  -- OK for SL --
   MIU time from 2024-01-21T08:25:49.518.648.000 to 2024-01-21T08:25:49.518.937.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.919084

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.2531974 to 42.2534864
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-21T08:26:11.438.021.000 to 2024-01-21T08:26:11.438.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.784477400 corrected by 40bit 10MHz counter of 7.815294300 obtained at 1Hz counter of 6. Result: 6.969186395
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-21T08:25:37.969.042.725
   Max 1st sample MIU time among all SL packets: 2024-01-21T08:26:11.438.169.725
   Min 1st sample XO time among all SL packets: 6.969046120
   Max 1st sample XO time among all SL packets: 40.438171359
   TM values from -269 to 276 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-21T08:25:37.969.042.725 (XO: 6.969046120 - 6.969352121, t=32.4 degC)  Rank: 00080
   TM values: Max =   236  Min =  -243  Ave =    -4.49  Std =    24.31
   Min TM value:-243 at an offset of 126.188 usec -> 2024-01-21T08:25:37.969.168.912 (XO: 6.969172308)
   Max TM value:236 at an offset of 110.463 usec -> 2024-01-21T08:25:37.969.153.187 (XO: 6.969156583)
   Max dTM/dt [1/us]:2.46e+03 at an offset of 126.263 usec -> 2024-01-21T08:25:37.969.168.987 (XO: 6.969172383)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 110.575 usec -> 2024-01-21T08:25:37.969.153.300 (XO: 6.969156695)

 - Block number: 2 starting at 2024-01-21T08:25:49.518.507.725 (XO: 18.518510240 - 18.518816241, t=34.0 degC)  Rank: 00077
   TM values: Max =   254  Min =  -265  Ave =    -4.59  Std =    22.86
   Min TM value:-265 at an offset of 123.650 usec -> 2024-01-21T08:25:49.518.631.375 (XO: 18.518633890)
   Max TM value:254 at an offset of 123.388 usec -> 2024-01-21T08:25:49.518.631.112 (XO: 18.518633628)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 121.163 usec -> 2024-01-21T08:25:49.518.628.887 (XO: 18.518631403)
   Min dTM/dt [1/us]:-2.90e+03 at an offset of 123.500 usec -> 2024-01-21T08:25:49.518.631.225 (XO: 18.518633740)

 - Block number: 3 starting at 2024-01-21T08:26:11.437.880.725 (XO: 40.437882358 - 40.438188359, t=32.4 degC)  Rank: 00082
   TM values: Max =   276  Min =  -269  Ave =    -4.63  Std =    18.93
   Min TM value:-269 at an offset of 123.488 usec -> 2024-01-21T08:26:11.438.004.212 (XO: 40.438005846)
   Max TM value:276 at an offset of 123.238 usec -> 2024-01-21T08:26:11.438.003.962 (XO: 40.438005596)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 123.175 usec -> 2024-01-21T08:26:11.438.003.900 (XO: 40.438005533)
   Min dTM/dt [1/us]:-2.86e+03 at an offset of 123.350 usec -> 2024-01-21T08:26:11.438.004.075 (XO: 40.438005708)
