-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb  4 21:53:05 2024 UTC
-- File 20240121_051729_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.248126 to 25.638706
-each separately from 10s to 25s and from 248126us to 638706us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.390002 in 1 packets: 18
MIU packet time from 2024-01-21T05:17:42.248.126.000 to 2024-01-21T05:17:57.638.706.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.390002 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9098448 to 26.9097932
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999484 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240121_051729_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1579708 to 12.1582598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3899510  -- OK for SL --
   MIU time from 2024-01-21T05:17:42.248.126.000 to 2024-01-21T05:17:42.248.415.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.390002

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.5482108 to 27.5484998
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-21T05:17:57.638.417.000 to 2024-01-21T05:17:57.638.706.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.157970800 corrected by 40bit 10MHz counter of 11.909844800 obtained at 1Hz counter of 10. Result: 10.248126844
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-21T05:17:42.247.985.725
   Max 1st sample MIU time among all SL packets: 2024-01-21T05:17:57.638.565.725
   Min 1st sample XO time among all SL packets: 10.247986568
   Max 1st sample XO time among all SL packets: 25.638568496
   TM values from -239 to 265 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-21T05:17:42.247.985.725 (XO: 10.247986568 - 10.248292569, t=32.4 degC)  Rank: 00080
   TM values: Max =   265  Min =  -216  Ave =    -4.50  Std =    18.50
   Min TM value:-216 at an offset of 130.938 usec -> 2024-01-21T05:17:42.248.116.662 (XO: 10.248117506)
   Max TM value:265 at an offset of 133.338 usec -> 2024-01-21T05:17:42.248.119.062 (XO: 10.248119906)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 133.250 usec -> 2024-01-21T05:17:42.248.118.975 (XO: 10.248119819)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 133.438 usec -> 2024-01-21T05:17:42.248.119.162 (XO: 10.248120006)

 - Block number: 2 starting at 2024-01-21T05:17:57.638.276.725 (XO: 25.638279495 - 25.638585496, t=32.4 degC)  Rank: 00078
   TM values: Max =   253  Min =  -239  Ave =    -4.53  Std =    17.90
   Min TM value:-239 at an offset of 132.050 usec -> 2024-01-21T05:17:57.638.408.775 (XO: 25.638411546)
   Max TM value:253 at an offset of 131.800 usec -> 2024-01-21T05:17:57.638.408.525 (XO: 25.638411296)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 129.475 usec -> 2024-01-21T05:17:57.638.406.200 (XO: 25.638408971)
   Min dTM/dt [1/us]:-2.56e+03 at an offset of 131.913 usec -> 2024-01-21T05:17:57.638.408.637 (XO: 25.638411408)
