-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb  4 21:21:54 2024 UTC
-- File 20240121_042809_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.578194 to 17.980082
-each separately from 8s to 17s and from 578194us to 980082us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.401310 in 1 packets: 18
MIU packet time from 2024-01-21T04:28:20.578.194.000 to 2024-01-21T04:28:29.980.082.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.401310 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4638070 to 18.4637761
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999691 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 36 packets (all packets)
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240121_042809_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0420012 to 10.0422902
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4012790  -- OK for SL --
   MIU time from 2024-01-21T04:28:20.578.194.000 to 2024-01-21T04:28:20.578.483.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.401310

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4435692 to 19.4438582
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-21T04:28:29.979.793.000 to 2024-01-21T04:28:29.980.082.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.042001200 corrected by 40bit 10MHz counter of 9.463807000 obtained at 1Hz counter of 8. Result: 8.578196224
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-21T04:28:20.578.053.725
   Max 1st sample MIU time among all SL packets: 2024-01-21T04:28:29.979.941.725
   Min 1st sample XO time among all SL packets: 8.578055948
   Max 1st sample XO time among all SL packets: 17.979945157
   TM values from -253 to 266 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-21T04:28:20.578.053.725 (XO: 8.578055948 - 8.578361949, t=32.4 degC)  Rank: 00079
   TM values: Max =   220  Min =  -253  Ave =    -4.40  Std =    20.77
   Min TM value:-253 at an offset of 129.613 usec -> 2024-01-21T04:28:20.578.183.337 (XO: 8.578185561)
   Max TM value:220 at an offset of 129.375 usec -> 2024-01-21T04:28:20.578.183.100 (XO: 8.578185324)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 127.188 usec -> 2024-01-21T04:28:20.578.180.912 (XO: 8.578183136)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 129.513 usec -> 2024-01-21T04:28:20.578.183.237 (XO: 8.578185461)

 - Block number: 2 starting at 2024-01-21T04:28:29.979.652.725 (XO: 17.979656156 - 17.979962157, t=32.4 degC)  Rank: 00086
   TM values: Max =   266  Min =  -230  Ave =    -4.44  Std =    19.93
   Min TM value:-230 at an offset of 129.775 usec -> 2024-01-21T04:28:29.979.782.500 (XO: 17.979785931)
   Max TM value:266 at an offset of 129.550 usec -> 2024-01-21T04:28:29.979.782.275 (XO: 17.979785706)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 132.388 usec -> 2024-01-21T04:28:29.979.785.112 (XO: 17.979788544)
   Min dTM/dt [1/us]:-2.85e+03 at an offset of 129.650 usec -> 2024-01-21T04:28:29.979.782.375 (XO: 17.979785806)
