-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Feb  4 21:08:54 2024 UTC
-- File 20240121_035953_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.908840 to 18.849222
-each separately from 14s to 18s and from 848933us to 909129us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.939804 in 1 packets: 18
MIU packet time from 2024-01-21T04:00:10.908.840.000 to 2024-01-21T04:00:14.849.222.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.939804 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.9117457 to 19.9117319
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999862 in 1 packets: 18
Temperature FPGA: 
      08 -> 44.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      09 -> 43.3 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      BE -> 32.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20240121_035953_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8205852 to 16.8208742
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.9397910  -- OK for SL --
   MIU time from 2024-01-21T04:00:10.908.840.000 to 2024-01-21T04:00:10.909.129.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.939804

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7606652 to 20.7609542
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2024-01-21T04:00:14.848.933.000 to 2024-01-21T04:00:14.849.222.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.820585200 corrected by 40bit 10MHz counter of 15.911745700 obtained at 1Hz counter of 14. Result: 14.908842681
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2024-01-21T04:00:10.908.699.725
   Max 1st sample MIU time among all SL packets: 2024-01-21T04:00:14.849.081.725
   Min 1st sample XO time among all SL packets: 14.908702405
   Max 1st sample XO time among all SL packets: 18.849084912
   TM values from -226 to 271 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2024-01-21T04:00:10.908.699.725 (XO: 14.908702405 - 14.909008407, t=32.4 degC)  Rank: 00080
   TM values: Max =   271  Min =  -222  Ave =    -4.48  Std =    18.26
   Min TM value:-222 at an offset of 95.350 usec -> 2024-01-21T04:00:10.908.795.075 (XO: 14.908797756)
   Max TM value:271 at an offset of 92.575 usec -> 2024-01-21T04:00:10.908.792.300 (XO: 14.908794981)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 95.450 usec -> 2024-01-21T04:00:10.908.795.175 (XO: 14.908797856)
   Min dTM/dt [1/us]:-2.66e+03 at an offset of 92.675 usec -> 2024-01-21T04:00:10.908.792.400 (XO: 14.908795081)

 - Block number: 2 starting at 2024-01-21T04:00:14.848.792.725 (XO: 18.848795911 - 18.849101912, t=32.4 degC)  Rank: 00075
   TM values: Max =   203  Min =  -226  Ave =    -4.53  Std =    21.02
   Min TM value:-226 at an offset of 130.850 usec -> 2024-01-21T04:00:14.848.923.575 (XO: 18.848926761)
   Max TM value:203 at an offset of 128.063 usec -> 2024-01-21T04:00:14.848.920.787 (XO: 18.848923974)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 130.950 usec -> 2024-01-21T04:00:14.848.923.675 (XO: 18.848926861)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 128.188 usec -> 2024-01-21T04:00:14.848.920.912 (XO: 18.848924099)
