-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Feb  1 03:04:22 2024 UTC
-- File 20231027_210458_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.833841 to 26.463295
-each separately from 8s to 26s and from 463006us to 834130us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.628876 in 1 packets: 18
MIU packet time from 2023-10-27T21:05:09.833.841.000 to 2023-10-27T21:05:27.463.295.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.628876 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8423693 to 27.8423078
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999385 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231027_210458_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6762111 to 10.6765001
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6288150  -- OK for SL --
   MIU time from 2023-10-27T21:05:09.833.841.000 to 2023-10-27T21:05:09.834.130.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.628876

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3053151 to 28.3056041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-27T21:05:27.463.006.000 to 2023-10-27T21:05:27.463.295.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.676211100 corrected by 40bit 10MHz counter of 9.842369300 obtained at 1Hz counter of 8. Result: 8.833844635
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-27T21:05:09.833.700.725
   Max 1st sample MIU time among all SL packets: 2023-10-27T21:05:27.463.154.725
   Min 1st sample XO time among all SL packets: 8.833704360
   Max 1st sample XO time among all SL packets: 26.463157600
   TM values from -253 to 207 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-27T21:05:09.833.700.725 (XO: 8.833704360 - 8.834010361, t=37.2 degC)  Rank: 00076
   TM values: Max =   207  Min =  -253  Ave =    -4.43  Std =    19.02
   Min TM value:-253 at an offset of 119.400 usec -> 2023-10-27T21:05:09.833.820.125 (XO: 8.833823760)
   Max TM value:207 at an offset of 119.088 usec -> 2023-10-27T21:05:09.833.819.812 (XO: 8.833823447)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 121.938 usec -> 2023-10-27T21:05:09.833.822.662 (XO: 8.833826298)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 119.213 usec -> 2023-10-27T21:05:09.833.819.937 (XO: 8.833823573)

 - Block number: 2 starting at 2023-10-27T21:05:27.462.865.725 (XO: 26.462868599 - 26.463174600, t=37.2 degC)  Rank: 00086
   TM values: Max =   199  Min =  -237  Ave =    -4.52  Std =    21.83
   Min TM value:-237 at an offset of 130.263 usec -> 2023-10-27T21:05:27.462.995.987 (XO: 26.462998862)
   Max TM value:199 at an offset of 125.900 usec -> 2023-10-27T21:05:27.462.991.625 (XO: 26.462994499)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 130.400 usec -> 2023-10-27T21:05:27.462.996.125 (XO: 26.462998999)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 114.888 usec -> 2023-10-27T21:05:27.462.980.612 (XO: 26.462983487)
