-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 23:15:51 2024 UTC
-- File 20231027_062247_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.595515 to 92.010787
-each separately from 1s to 92s and from 10498us to 867891us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.150490 in 1 packets: 18
      14.730435 in 1 packets: 36
      7.390295 in 1 packets: 54
      25.481175 in 1 packets: 72
      6.850295 in 1 packets: 90
      13.810559 in 1 packets: 108
MIU packet time from 2023-10-27T06:22:51.595.515.000 to 2023-10-27T06:24:22.010.787.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.150490 in 1 packets: 18
      14.730435 in 1 packets: 36
      7.390295 in 1 packets: 54
      25.481175 in 1 packets: 72
      6.850295 in 1 packets: 90
      13.810559 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8554378 to 93.8551271
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999249 in 1 packets: 18
      14.9999488 in 1 packets: 36
      6.9999760 in 1 packets: 54
      25.9999113 in 1 packets: 72
      6.9999761 in 1 packets: 90
      13.9999522 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 126 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20231027_062247_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.4509540 to 3.4512430
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1504150  -- OK for SL --
   MIU time from 2023-10-27T06:22:51.595.515.000 to 2023-10-27T06:22:51.595.804.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.150490

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.6016580 to 25.6019470
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7303830  -- OK for SL --
   MIU time from 2023-10-27T06:23:13.746.294.000 to 2023-10-27T06:23:13.746.583.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.730435

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      99 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.3323300 to 40.3326190
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.3902710  -- OK for SL --
   MIU time from 2023-10-27T06:23:28.477.018.000 to 2023-10-27T06:23:28.477.307.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.390295

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.7228900 to 47.7231790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.4810870  -- OK for SL --
   MIU time from 2023-10-27T06:23:35.867.602.000 to 2023-10-27T06:23:35.867.891.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.481175

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 73.2042660 to 73.2045550
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8502710  -- OK for SL --
   MIU time from 2023-10-27T06:24:01.349.066.000 to 2023-10-27T06:24:01.349.355.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.850295

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      104 in 18 packets (all packets)  -- OK --
   10 MHz counter from 80.0548260 to 80.0551150
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8105110  -- OK for SL --
   MIU time from 2023-10-27T06:24:08.199.650.000 to 2023-10-27T06:24:08.199.939.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.810559

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 93.8656260 to 93.8659150
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-27T06:24:22.010.498.000 to 2023-10-27T06:24:22.010.787.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.450954000 corrected by 40bit 10MHz counter of 2.855437800 obtained at 1Hz counter of 1. Result: 1.595518225
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-27T06:22:51.595.374.725
   Max 1st sample MIU time among all SL packets: 2023-10-27T06:24:22.010.646.725
   Min 1st sample XO time among all SL packets: 1.595377949
   Max 1st sample XO time among all SL packets: 92.010647661
   TM values from -844 to 1042 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-27T06:22:51.595.374.725 (XO: 1.595377949 - 1.595683950, t=37.2 degC)  Rank: 00077
   TM values: Max =   820  Min =  -820  Ave =    -4.61  Std =    38.96
   Min TM value:-820 at an offset of 90.500 usec -> 2023-10-27T06:22:51.595.465.225 (XO: 1.595468450)
   Max TM value:820 at an offset of 90.300 usec -> 2023-10-27T06:22:51.595.465.025 (XO: 1.595468250)
   Max dTM/dt [1/us]:1.29e+04 at an offset of 90.275 usec -> 2023-10-27T06:22:51.595.465.000 (XO: 1.595468225)
   Min dTM/dt [1/us]:-8.87e+03 at an offset of 90.463 usec -> 2023-10-27T06:22:51.595.465.187 (XO: 1.595468412)

 - Block number: 2 starting at 2023-10-27T06:23:13.746.153.725 (XO: 23.746157562 - 23.746463563, t=37.2 degC)  Rank: 00073
   TM values: Max =   177  Min =  -247  Ave =    -4.52  Std =    17.30
   Min TM value:-247 at an offset of 114.300 usec -> 2023-10-27T06:23:13.746.268.025 (XO: 23.746271862)
   Max TM value:177 at an offset of 113.913 usec -> 2023-10-27T06:23:13.746.267.637 (XO: 23.746271475)
   Max dTM/dt [1/us]:1.26e+03 at an offset of 116.863 usec -> 2023-10-27T06:23:13.746.270.587 (XO: 23.746274425)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 101.588 usec -> 2023-10-27T06:23:13.746.255.312 (XO: 23.746259150)

 - Block number: 3 starting at 2023-10-27T06:23:28.476.877.725 (XO: 38.476879894 - 38.477185895, t=37.2 degC)  Rank: 00099
   TM values: Max =   977  Min =  -804  Ave =    -4.48  Std =    30.06
   Min TM value:-804 at an offset of 131.225 usec -> 2023-10-27T06:23:28.477.008.950 (XO: 38.477011120)
   Max TM value:977 at an offset of 131.025 usec -> 2023-10-27T06:23:28.477.008.750 (XO: 38.477010920)
   Max dTM/dt [1/us]:1.38e+04 at an offset of 131.000 usec -> 2023-10-27T06:23:28.477.008.725 (XO: 38.477010895)
   Min dTM/dt [1/us]:-8.95e+03 at an offset of 131.188 usec -> 2023-10-27T06:23:28.477.008.912 (XO: 38.477011082)

 - Block number: 4 starting at 2023-10-27T06:23:35.867.461.725 (XO: 45.867465088 - 45.867771089, t=37.2 degC)  Rank: 00068
   TM values: Max =   220  Min =  -236  Ave =    -4.45  Std =    19.48
   Min TM value:-236 at an offset of 125.825 usec -> 2023-10-27T06:23:35.867.587.550 (XO: 45.867590913)
   Max TM value:220 at an offset of 125.538 usec -> 2023-10-27T06:23:35.867.587.262 (XO: 45.867590626)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 128.250 usec -> 2023-10-27T06:23:35.867.589.975 (XO: 45.867593338)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 125.650 usec -> 2023-10-27T06:23:35.867.587.375 (XO: 45.867590738)

 - Block number: 5 starting at 2023-10-27T06:24:01.348.925.725 (XO: 71.348928111 - 71.349234112, t=37.2 degC)  Rank: 00073
   TM values: Max =   174  Min =  -220  Ave =    -4.48  Std =    17.83
   Min TM value:-220 at an offset of 125.738 usec -> 2023-10-27T06:24:01.349.051.462 (XO: 71.349053849)
   Max TM value:174 at an offset of 113.613 usec -> 2023-10-27T06:24:01.349.039.337 (XO: 71.349041724)
   Max dTM/dt [1/us]:1.30e+03 at an offset of 128.325 usec -> 2023-10-27T06:24:01.349.054.050 (XO: 71.349056437)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 113.700 usec -> 2023-10-27T06:24:01.349.039.425 (XO: 71.349041812)

 - Block number: 6 starting at 2023-10-27T06:24:08.199.509.725 (XO: 78.199511503 - 78.199817504, t=37.2 degC)  Rank: 00104
   TM values: Max =  1042  Min =  -844  Ave =    -4.44  Std =    34.15
   Min TM value:-844 at an offset of 131.313 usec -> 2023-10-27T06:24:08.199.641.037 (XO: 78.199642816)
   Max TM value:1042 at an offset of 131.113 usec -> 2023-10-27T06:24:08.199.640.837 (XO: 78.199642616)
   Max dTM/dt [1/us]:1.36e+04 at an offset of 131.088 usec -> 2023-10-27T06:24:08.199.640.812 (XO: 78.199642591)
   Min dTM/dt [1/us]:-8.94e+03 at an offset of 131.275 usec -> 2023-10-27T06:24:08.199.641.000 (XO: 78.199642779)

 - Block number: 7 starting at 2023-10-27T06:24:22.010.357.725 (XO: 92.010358660 - 92.010664661, t=37.2 degC)  Rank: 00069
   TM values: Max =   199  Min =  -248  Ave =    -4.45  Std =    21.69
   Min TM value:-248 at an offset of 112.600 usec -> 2023-10-27T06:24:22.010.470.325 (XO: 92.010471261)
   Max TM value:199 at an offset of 112.250 usec -> 2023-10-27T06:24:22.010.469.975 (XO: 92.010470911)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 115.050 usec -> 2023-10-27T06:24:22.010.472.775 (XO: 92.010473711)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 112.375 usec -> 2023-10-27T06:24:22.010.470.100 (XO: 92.010471036)
