-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 17:38:55 2024 UTC
-- File 20231026_124804_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.313451 to 85.146187
-each separately from 6s to 85s and from 145898us to 775993us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.270485 in 1 packets: 18
      22.040763 in 1 packets: 36
      8.150138 in 1 packets: 54
      19.689620 in 1 packets: 72
      17.679996 in 1 packets: 90
MIU packet time from 2023-10-26T12:48:13.313.451.000 to 2023-10-26T12:49:32.146.187.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.270485 in 1 packets: 18
      22.040763 in 1 packets: 36
      8.150138 in 1 packets: 54
      19.689620 in 1 packets: 72
      17.679996 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9150558 to 86.9147845
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999623 in 1 packets: 18
      21.9999245 in 1 packets: 36
      7.9999725 in 1 packets: 54
      19.9999314 in 1 packets: 72
      17.9999380 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      06 -> 47.2 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20231026_124804_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2285081 to 8.2287971
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2704470  -- OK for SL --
   MIU time from 2023-10-26T12:48:13.313.451.000 to 2023-10-26T12:48:13.313.740.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.270485

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4992441 to 19.4995331
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.0406870  -- OK for SL --
   MIU time from 2023-10-26T12:48:24.584.225.000 to 2023-10-26T12:48:24.584.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.040763

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.5402201 to 41.5405091
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1501110  -- OK for SL --
   MIU time from 2023-10-26T12:48:46.625.277.000 to 2023-10-26T12:48:46.625.566.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.150138

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.6906201 to 49.6909091
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6895510  -- OK for SL --
   MIU time from 2023-10-26T12:48:54.775.704.000 to 2023-10-26T12:48:54.775.993.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.689620

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.3804601 to 69.3807491
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6799350  -- OK for SL --
   MIU time from 2023-10-26T12:49:14.465.613.000 to 2023-10-26T12:49:14.465.902.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.679996

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 87.0606841 to 87.0609731
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-26T12:49:32.145.898.000 to 2023-10-26T12:49:32.146.187.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.228508100 corrected by 40bit 10MHz counter of 7.915055800 obtained at 1Hz counter of 6. Result: 6.313453366
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-26T12:48:13.313.310.725
   Max 1st sample MIU time among all SL packets: 2023-10-26T12:49:32.146.046.725
   Min 1st sample XO time among all SL packets: 6.313313090
   Max 1st sample XO time among all SL packets: 85.146048822
   TM values from -268 to 236 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-26T12:48:13.313.310.725 (XO: 6.313313090 - 6.313619091, t=37.2 degC)  Rank: 00078
   TM values: Max =   227  Min =  -214  Ave =    -4.59  Std =    22.68
   Min TM value:-214 at an offset of 111.788 usec -> 2023-10-26T12:48:13.313.422.512 (XO: 6.313424878)
   Max TM value:227 at an offset of 109.125 usec -> 2023-10-26T12:48:13.313.419.850 (XO: 6.313422216)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 111.888 usec -> 2023-10-26T12:48:13.313.422.612 (XO: 6.313424978)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 109.250 usec -> 2023-10-26T12:48:13.313.419.975 (XO: 6.313422341)

 - Block number: 2 starting at 2023-10-26T12:48:24.584.084.725 (XO: 17.584087769 - 17.584393770, t=37.2 degC)  Rank: 00077
   TM values: Max =   236  Min =  -246  Ave =    -4.55  Std =    18.97
   Min TM value:-246 at an offset of 107.725 usec -> 2023-10-26T12:48:24.584.192.450 (XO: 17.584195495)
   Max TM value:236 at an offset of 95.438 usec -> 2023-10-26T12:48:24.584.180.162 (XO: 17.584183207)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 110.275 usec -> 2023-10-26T12:48:24.584.195.000 (XO: 17.584198045)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 95.550 usec -> 2023-10-26T12:48:24.584.180.275 (XO: 17.584183320)

 - Block number: 3 starting at 2023-10-26T12:48:46.625.136.725 (XO: 39.625139350 - 39.625445352, t=37.2 degC)  Rank: 00082
   TM values: Max =   229  Min =  -268  Ave =    -4.53  Std =    19.19
   Min TM value:-268 at an offset of 122.113 usec -> 2023-10-26T12:48:46.625.258.837 (XO: 39.625261463)
   Max TM value:229 at an offset of 121.788 usec -> 2023-10-26T12:48:46.625.258.512 (XO: 39.625261138)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 119.675 usec -> 2023-10-26T12:48:46.625.256.400 (XO: 39.625259026)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 121.913 usec -> 2023-10-26T12:48:46.625.258.637 (XO: 39.625261263)

 - Block number: 4 starting at 2023-10-26T12:48:54.775.563.725 (XO: 47.775567362 - 47.775873363, t=37.2 degC)  Rank: 00077
   TM values: Max =   190  Min =  -225  Ave =    -4.51  Std =    19.19
   Min TM value:-225 at an offset of 123.900 usec -> 2023-10-26T12:48:54.775.687.625 (XO: 47.775691262)
   Max TM value:190 at an offset of 121.163 usec -> 2023-10-26T12:48:54.775.684.887 (XO: 47.775688525)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 124.000 usec -> 2023-10-26T12:48:54.775.687.725 (XO: 47.775691362)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 121.300 usec -> 2023-10-26T12:48:54.775.685.025 (XO: 47.775688662)

 - Block number: 5 starting at 2023-10-26T12:49:14.465.472.725 (XO: 67.465474954 - 67.465780955, t=37.2 degC)  Rank: 00071
   TM values: Max =   220  Min =  -254  Ave =    -4.50  Std =    19.93
   Min TM value:-254 at an offset of 120.238 usec -> 2023-10-26T12:49:14.465.592.962 (XO: 67.465595192)
   Max TM value:220 at an offset of 119.938 usec -> 2023-10-26T12:49:14.465.592.662 (XO: 67.465594892)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 122.713 usec -> 2023-10-26T12:49:14.465.595.437 (XO: 67.465597667)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 120.063 usec -> 2023-10-26T12:49:14.465.592.787 (XO: 67.465595017)

 - Block number: 6 starting at 2023-10-26T12:49:32.145.757.725 (XO: 85.145759821 - 85.146065822, t=37.2 degC)  Rank: 00086
   TM values: Max =   218  Min =  -250  Ave =    -4.50  Std =    20.97
   Min TM value:-250 at an offset of 106.675 usec -> 2023-10-26T12:49:32.145.864.400 (XO: 85.145866496)
   Max TM value:218 at an offset of 106.375 usec -> 2023-10-26T12:49:32.145.864.100 (XO: 85.145866196)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 121.738 usec -> 2023-10-26T12:49:32.145.879.462 (XO: 85.145881558)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 106.500 usec -> 2023-10-26T12:49:32.145.864.225 (XO: 85.145866321)
