-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 16:11:13 2024 UTC
-- File 20231026_102526_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.840635 to 82.390938
-each separately from 3s to 82s and from 346303us to 840924us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.892574 in 1 packets: 18
      14.081423 in 1 packets: 36
      10.530804 in 1 packets: 54
      17.382345 in 1 packets: 72
      13.661423 in 1 packets: 90
MIU packet time from 2023-10-26T10:25:32.840.635.000 to 2023-10-26T10:26:51.390.938.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.892574 in 1 packets: 18
      14.081423 in 1 packets: 36
      10.530804 in 1 packets: 54
      17.382345 in 1 packets: 72
      13.661423 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8652032 to 83.8649326
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999208 in 1 packets: 18
      13.9999519 in 1 packets: 36
      10.9999624 in 1 packets: 54
      16.9999421 in 1 packets: 72
      13.9999522 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      03 -> 51.1 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      2 in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20231026_102526_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.7058386 to 5.7061276
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.8924950  -- OK for SL --
   MIU time from 2023-10-26T10:25:32.840.635.000 to 2023-10-26T10:25:32.840.924.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.892574

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.5986226 to 28.5989116
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0813750  -- OK for SL --
   MIU time from 2023-10-26T10:25:55.733.498.000 to 2023-10-26T10:25:55.733.787.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.081423

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.6802866 to 42.6805756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5307670  -- OK for SL --
   MIU time from 2023-10-26T10:26:09.815.210.000 to 2023-10-26T10:26:09.815.499.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.530804

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.2113426 to 53.2116316
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3822870  -- OK for SL --
   MIU time from 2023-10-26T10:26:20.346.303.000 to 2023-10-26T10:26:20.346.592.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.382345

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.5939186 to 70.5942076
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6613750  -- OK for SL --
   MIU time from 2023-10-26T10:26:37.728.937.000 to 2023-10-26T10:26:37.729.226.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.661423

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 84.2555826 to 84.2558716
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-26T10:26:51.390.649.000 to 2023-10-26T10:26:51.390.938.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.705838600 corrected by 40bit 10MHz counter of 4.865203200 obtained at 1Hz counter of 3. Result: 3.840638342
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-26T10:25:32.840.494.725
   Max 1st sample MIU time among all SL packets: 2023-10-26T10:26:51.390.797.725
   Min 1st sample XO time among all SL packets: 3.840498067
   Max 1st sample XO time among all SL packets: 82.390800054
   TM values from -251 to 234 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-26T10:25:32.840.494.725 (XO: 3.840498067 - 3.840804068, t=37.2 degC)  Rank: 00077
   TM values: Max =   234  Min =  -251  Ave =    -4.48  Std =    20.01
   Min TM value:-251 at an offset of 124.263 usec -> 2023-10-26T10:25:32.840.618.987 (XO: 3.840622330)
   Max TM value:234 at an offset of 112.163 usec -> 2023-10-26T10:25:32.840.606.887 (XO: 3.840610230)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 126.788 usec -> 2023-10-26T10:25:32.840.621.512 (XO: 3.840624855)
   Min dTM/dt [1/us]:-2.33e+03 at an offset of 112.263 usec -> 2023-10-26T10:25:32.840.606.987 (XO: 3.840610330)

 - Block number: 2 starting at 2023-10-26T10:25:55.733.357.725 (XO: 26.733360818 - 26.733666819, t=37.2 degC)  Rank: 00079
   TM values: Max =   171  Min =  -234  Ave =    -4.65  Std =    20.23
   Min TM value:-234 at an offset of 112.450 usec -> 2023-10-26T10:25:55.733.470.175 (XO: 26.733473269)
   Max TM value:171 at an offset of 112.088 usec -> 2023-10-26T10:25:55.733.469.812 (XO: 26.733472906)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 115.025 usec -> 2023-10-26T10:25:55.733.472.750 (XO: 26.733475844)
   Min dTM/dt [1/us]:-1.44e+03 at an offset of 99.725 usec -> 2023-10-26T10:25:55.733.457.450 (XO: 26.733460544)

 - Block number: 3 starting at 2023-10-26T10:26:09.815.069.725 (XO: 40.815073196 - 40.815379197, t=37.2 degC)  Rank: 00078
   TM values: Max =   192  Min =  -220  Ave =    -4.52  Std =    17.48
   Min TM value:-220 at an offset of 126.963 usec -> 2023-10-26T10:26:09.815.196.687 (XO: 40.815200159)
   Max TM value:192 at an offset of 127.188 usec -> 2023-10-26T10:26:09.815.196.912 (XO: 40.815200384)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 127.063 usec -> 2023-10-26T10:26:09.815.196.787 (XO: 40.815200259)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 126.875 usec -> 2023-10-26T10:26:09.815.196.600 (XO: 40.815200072)

 - Block number: 4 starting at 2023-10-26T10:26:20.346.162.725 (XO: 51.346165202 - 51.346471203, t=37.2 degC)  Rank: 00084
   TM values: Max =   225  Min =  -250  Ave =    -4.61  Std =    25.97
   Min TM value:-250 at an offset of 156.651 usec -> 2023-10-26T10:26:20.346.319.375 (XO: 51.346321852)
   Max TM value:225 at an offset of 156.388 usec -> 2023-10-26T10:26:20.346.319.113 (XO: 51.346321590)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 100.338 usec -> 2023-10-26T10:26:20.346.263.062 (XO: 51.346265540)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 156.501 usec -> 2023-10-26T10:26:20.346.319.225 (XO: 51.346321702)

 - Block number: 5 starting at 2023-10-26T10:26:37.728.796.725 (XO: 68.728800403 - 68.729106404, t=37.2 degC)  Rank: 00079
   TM values: Max =   191  Min =  -233  Ave =    -4.43  Std =    19.10
   Min TM value:-233 at an offset of 125.000 usec -> 2023-10-26T10:26:37.728.921.725 (XO: 68.728925403)
   Max TM value:191 at an offset of 124.650 usec -> 2023-10-26T10:26:37.728.921.375 (XO: 68.728925053)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 127.575 usec -> 2023-10-26T10:26:37.728.924.300 (XO: 68.728927978)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 124.850 usec -> 2023-10-26T10:26:37.728.921.575 (XO: 68.728925253)

 - Block number: 6 starting at 2023-10-26T10:26:51.390.508.725 (XO: 82.390511053 - 82.390817054, t=37.2 degC)  Rank: 00079
   TM values: Max =   196  Min =  -239  Ave =    -4.55  Std =    17.06
   Min TM value:-239 at an offset of 127.563 usec -> 2023-10-26T10:26:51.390.636.287 (XO: 82.390638616)
   Max TM value:196 at an offset of 127.200 usec -> 2023-10-26T10:26:51.390.635.925 (XO: 82.390638253)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 130.125 usec -> 2023-10-26T10:26:51.390.638.850 (XO: 82.390641178)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 115.413 usec -> 2023-10-26T10:26:51.390.624.137 (XO: 82.390626466)
