-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 15:36:24 2024 UTC
-- File 20231026_093405_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.007064 to 37.739111
-each separately from 14s to 37s and from 7064us to 977767us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.970125 in 1 packets: 18
      18.761055 in 1 packets: 36
MIU packet time from 2023-10-26T09:34:22.007.064.000 to 2023-10-26T09:34:45.739.111.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.970125 in 1 packets: 18
      18.761055 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7873174 to 38.7872395
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999865 in 1 packets: 18
      18.9999356 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231026_093405_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.7943823 to 15.7946713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.9701110  -- OK for SL --
   MIU time from 2023-10-26T09:34:22.007.064.000 to 2023-10-26T09:34:22.007.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.970125

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.7647823 to 20.7650713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7609910  -- OK for SL --
   MIU time from 2023-10-26T09:34:26.977.478.000 to 2023-10-26T09:34:26.977.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.761055

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.5260623 to 39.5263513
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-26T09:34:45.738.822.000 to 2023-10-26T09:34:45.739.111.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.794382300 corrected by 40bit 10MHz counter of 15.787317400 obtained at 1Hz counter of 14. Result: 14.007064924
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-26T09:34:22.006.923.725
   Max 1st sample MIU time among all SL packets: 2023-10-26T09:34:45.738.970.725
   Min 1st sample XO time among all SL packets: 14.006924649
   Max 1st sample XO time among all SL packets: 37.738974038
   TM values from -248 to 219 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-26T09:34:22.006.923.725 (XO: 14.006924649 - 14.007230650, t=37.2 degC)  Rank: 00078
   TM values: Max =   174  Min =  -224  Ave =    -4.74  Std =    20.87
   Min TM value:-224 at an offset of 130.250 usec -> 2023-10-26T09:34:22.007.053.975 (XO: 14.007054899)
   Max TM value:174 at an offset of 115.250 usec -> 2023-10-26T09:34:22.007.038.975 (XO: 14.007039899)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 130.363 usec -> 2023-10-26T09:34:22.007.054.087 (XO: 14.007055011)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 115.388 usec -> 2023-10-26T09:34:22.007.039.112 (XO: 14.007040036)

 - Block number: 2 starting at 2023-10-26T09:34:26.977.337.725 (XO: 18.977341448 - 18.977647449, t=37.2 degC)  Rank: 00083
   TM values: Max =   219  Min =  -248  Ave =    -4.56  Std =    19.10
   Min TM value:-248 at an offset of 126.463 usec -> 2023-10-26T09:34:26.977.464.187 (XO: 18.977467911)
   Max TM value:219 at an offset of 126.125 usec -> 2023-10-26T09:34:26.977.463.850 (XO: 18.977467573)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 128.863 usec -> 2023-10-26T09:34:26.977.466.587 (XO: 18.977470311)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 126.238 usec -> 2023-10-26T09:34:26.977.463.962 (XO: 18.977467686)

 - Block number: 3 starting at 2023-10-26T09:34:45.738.681.725 (XO: 37.738685037 - 37.738991038, t=37.2 degC)  Rank: 00081
   TM values: Max =   211  Min =  -244  Ave =    -4.60  Std =    18.66
   Min TM value:-244 at an offset of 127.625 usec -> 2023-10-26T09:34:45.738.809.350 (XO: 37.738812662)
   Max TM value:211 at an offset of 127.350 usec -> 2023-10-26T09:34:45.738.809.075 (XO: 37.738812387)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 130.138 usec -> 2023-10-26T09:34:45.738.811.862 (XO: 37.738815174)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 127.463 usec -> 2023-10-26T09:34:45.738.809.187 (XO: 37.738812499)
