-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 31 12:46:39 2024 UTC
-- File 20231026_015232_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.757072 to 56.123033
-each separately from 2s to 56s and from 122744us to 757361us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.722188 in 1 packets: 18
      12.863611 in 1 packets: 36
      22.779006 in 1 packets: 54
MIU packet time from 2023-10-26T01:52:37.757.072.000 to 2023-10-26T01:53:31.123.033.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.722188 in 1 packets: 18
      12.863611 in 1 packets: 36
      22.779006 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9147856 to 57.9146013
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999385 in 1 packets: 18
      12.9999556 in 1 packets: 36
      22.9999216 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20231026_015232_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6718580 to 4.6721470
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.7221270  -- OK for SL --
   MIU time from 2023-10-26T01:52:37.757.072.000 to 2023-10-26T01:52:37.757.361.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.722188

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      106 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3942740 to 22.3945630
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8635670  -- OK for SL --
   MIU time from 2023-10-26T01:52:55.479.549.000 to 2023-10-26T01:52:55.479.838.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.863611

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.2581300 to 35.2584190
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.7789270  -- OK for SL --
   MIU time from 2023-10-26T01:53:08.343.449.000 to 2023-10-26T01:53:08.343.738.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.779006

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.0373460 to 58.0376350
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-26T01:53:31.122.744.000 to 2023-10-26T01:53:31.123.033.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.671858000 corrected by 40bit 10MHz counter of 3.914785600 obtained at 1Hz counter of 2. Result: 2.757075050
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-26T01:52:37.756.931.725
   Max 1st sample MIU time among all SL packets: 2023-10-26T01:53:31.122.892.725
   Min 1st sample XO time among all SL packets: 2.756934774
   Max 1st sample XO time among all SL packets: 56.122893855
   TM values from -986 to 1070 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-26T01:52:37.756.931.725 (XO: 2.756934774 - 2.757240775, t=37.2 degC)  Rank: 00075
   TM values: Max =   226  Min =  -246  Ave =    -4.51  Std =    20.91
   Min TM value:-246 at an offset of 128.188 usec -> 2023-10-26T01:52:37.757.059.912 (XO: 2.757062962)
   Max TM value:226 at an offset of 127.863 usec -> 2023-10-26T01:52:37.757.059.587 (XO: 2.757062637)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 130.563 usec -> 2023-10-26T01:52:37.757.062.287 (XO: 2.757065337)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 128.000 usec -> 2023-10-26T01:52:37.757.059.725 (XO: 2.757062775)

 - Block number: 2 starting at 2023-10-26T01:52:55.479.408.725 (XO: 20.479411303 - 20.479717304, t=37.2 degC)  Rank: 00106
   TM values: Max =  1070  Min =  -986  Ave =    -4.48  Std =    34.15
   Min TM value:-986 at an offset of 101.700 usec -> 2023-10-26T01:52:55.479.510.425 (XO: 20.479513003)
   Max TM value:1070 at an offset of 101.763 usec -> 2023-10-26T01:52:55.479.510.487 (XO: 20.479513066)
   Max dTM/dt [1/us]:1.48e+04 at an offset of 101.738 usec -> 2023-10-26T01:52:55.479.510.462 (XO: 20.479513041)
   Min dTM/dt [1/us]:-9.43e+03 at an offset of 101.925 usec -> 2023-10-26T01:52:55.479.510.650 (XO: 20.479513228)

 - Block number: 3 starting at 2023-10-26T01:53:08.343.308.725 (XO: 33.343311192 - 33.343617193, t=37.2 degC)  Rank: 00074
   TM values: Max =   227  Min =  -238  Ave =    -4.45  Std =    27.69
   Min TM value:-238 at an offset of 129.363 usec -> 2023-10-26T01:53:08.343.438.087 (XO: 33.343440555)
   Max TM value:227 at an offset of 129.025 usec -> 2023-10-26T01:53:08.343.437.750 (XO: 33.343440218)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 126.925 usec -> 2023-10-26T01:53:08.343.435.650 (XO: 33.343438118)
   Min dTM/dt [1/us]:-1.93e+03 at an offset of 129.100 usec -> 2023-10-26T01:53:08.343.437.825 (XO: 33.343440293)

 - Block number: 4 starting at 2023-10-26T01:53:31.122.603.725 (XO: 56.122604854 - 56.122910855, t=37.2 degC)  Rank: 00089
   TM values: Max =  1005  Min =  -891  Ave =    -4.55  Std =    36.35
   Min TM value:-891 at an offset of 115.438 usec -> 2023-10-26T01:53:31.122.719.162 (XO: 56.122720292)
   Max TM value:1005 at an offset of 115.500 usec -> 2023-10-26T01:53:31.122.719.225 (XO: 56.122720355)
   Max dTM/dt [1/us]:1.37e+04 at an offset of 115.475 usec -> 2023-10-26T01:53:31.122.719.200 (XO: 56.122720330)
   Min dTM/dt [1/us]:-9.13e+03 at an offset of 115.663 usec -> 2023-10-26T01:53:31.122.719.387 (XO: 56.122720517)
