-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 29 11:01:22 2024 UTC
-- File 20231020_140933_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.426901 to 46.578677
-each separately from 4s to 46s and from 217647us to 578677us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.790457 in 1 packets: 18
      18.110364 in 1 packets: 36
      7.249799 in 1 packets: 54
MIU packet time from 2023-10-20T14:09:40.426.901.000 to 2023-10-20T14:10:22.578.677.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.790457 in 1 packets: 18
      18.110364 in 1 packets: 36
      7.249799 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8908536 to 47.8907111
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999424 in 1 packets: 18
      17.9999389 in 1 packets: 36
      6.9999762 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20231020_140933_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.3177557 to 6.3180447
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7903990  -- OK for SL --
   MIU time from 2023-10-20T14:09:40.426.901.000 to 2023-10-20T14:09:40.427.190.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.790457

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.1084437 to 23.1087327
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.1103030  -- OK for SL --
   MIU time from 2023-10-20T14:09:57.217.647.000 to 2023-10-20T14:09:57.217.936.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.110364

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.2190357 to 41.2193247
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2497750  -- OK for SL --
   MIU time from 2023-10-20T14:10:15.328.300.000 to 2023-10-20T14:10:15.328.589.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.249799

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 48.4690997 to 48.4693887
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-20T14:10:22.578.388.000 to 2023-10-20T14:10:22.578.677.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.317755700 corrected by 40bit 10MHz counter of 5.890853600 obtained at 1Hz counter of 4. Result: 4.426903551
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-20T14:09:40.426.760.725
   Max 1st sample MIU time among all SL packets: 2023-10-20T14:10:22.578.536.725
   Min 1st sample XO time among all SL packets: 4.426763276
   Max 1st sample XO time among all SL packets: 46.578539292
   TM values from -251 to 200 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-20T14:09:40.426.760.725 (XO: 4.426763276 - 4.427069277, t=38.8 degC)  Rank: 00082
   TM values: Max =   184  Min =  -210  Ave =    -4.51  Std =    20.36
   Min TM value:-210 at an offset of 126.400 usec -> 2023-10-20T14:09:40.426.887.125 (XO: 4.426889676)
   Max TM value:184 at an offset of 126.088 usec -> 2023-10-20T14:09:40.426.886.812 (XO: 4.426889364)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 128.863 usec -> 2023-10-20T14:09:40.426.889.587 (XO: 4.426892139)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 126.213 usec -> 2023-10-20T14:09:40.426.886.937 (XO: 4.426889489)

 - Block number: 2 starting at 2023-10-20T14:09:57.217.506.725 (XO: 21.217508165 - 21.217814166, t=38.8 degC)  Rank: 00078
   TM values: Max =   178  Min =  -249  Ave =    -4.55  Std =    20.57
   Min TM value:-249 at an offset of 121.650 usec -> 2023-10-20T14:09:57.217.628.375 (XO: 21.217629815)
   Max TM value:178 at an offset of 108.988 usec -> 2023-10-20T14:09:57.217.615.712 (XO: 21.217617152)
   Max dTM/dt [1/us]:1.18e+03 at an offset of 124.275 usec -> 2023-10-20T14:09:57.217.631.000 (XO: 21.217632440)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 109.150 usec -> 2023-10-20T14:09:57.217.615.875 (XO: 21.217617315)

 - Block number: 3 starting at 2023-10-20T14:10:15.328.159.725 (XO: 39.328161641 - 39.328467642, t=38.8 degC)  Rank: 00082
   TM values: Max =   200  Min =  -217  Ave =    -4.53  Std =    18.67
   Min TM value:-217 at an offset of 126.425 usec -> 2023-10-20T14:10:15.328.286.150 (XO: 39.328288066)
   Max TM value:200 at an offset of 126.075 usec -> 2023-10-20T14:10:15.328.285.800 (XO: 39.328287716)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 128.825 usec -> 2023-10-20T14:10:15.328.288.550 (XO: 39.328290466)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 126.238 usec -> 2023-10-20T14:10:15.328.285.962 (XO: 39.328287879)

 - Block number: 4 starting at 2023-10-20T14:10:22.578.247.725 (XO: 46.578250291 - 46.578556292, t=38.8 degC)  Rank: 00080
   TM values: Max =   182  Min =  -251  Ave =    -4.60  Std =    23.63
   Min TM value:-251 at an offset of 123.663 usec -> 2023-10-20T14:10:22.578.371.387 (XO: 46.578373954)
   Max TM value:182 at an offset of 123.313 usec -> 2023-10-20T14:10:22.578.371.037 (XO: 46.578373604)
   Max dTM/dt [1/us]:1.38e+03 at an offset of 126.163 usec -> 2023-10-20T14:10:22.578.373.887 (XO: 46.578376454)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 111.513 usec -> 2023-10-20T14:10:22.578.359.237 (XO: 46.578361804)
