-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 22:33:02 2024 UTC
-- File 20231016_072201_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.117195 to 19.856894
-each separately from 9s to 19s and from 117195us to 856894us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.739121 in 1 packets: 18
MIU packet time from 2023-10-16T07:22:13.117.195.000 to 2023-10-16T07:22:23.856.894.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.739121 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9067542 to 20.9067200
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999658 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231016_072201_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.0239499 to 11.0242389
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.7390870  -- OK for SL --
   MIU time from 2023-10-16T07:22:13.117.195.000 to 2023-10-16T07:22:13.117.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.739121

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.7633259 to 21.7636149
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-16T07:22:23.856.605.000 to 2023-10-16T07:22:23.856.894.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.023949900 corrected by 40bit 10MHz counter of 10.906754200 obtained at 1Hz counter of 9. Result: 9.117196098
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-16T07:22:13.117.054.725
   Max 1st sample MIU time among all SL packets: 2023-10-16T07:22:23.856.753.725
   Min 1st sample XO time among all SL packets: 9.117055823
   Max 1st sample XO time among all SL packets: 19.856757538
   TM values from -286 to 227 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-16T07:22:13.117.054.725 (XO: 9.117055823 - 9.117361824, t=38.8 degC)  Rank: 00085
   TM values: Max =   227  Min =  -286  Ave =    -4.42  Std =    24.83
   Min TM value:-286 at an offset of 125.938 usec -> 2023-10-16T07:22:13.117.180.662 (XO: 9.117181761)
   Max TM value:227 at an offset of 125.638 usec -> 2023-10-16T07:22:13.117.180.362 (XO: 9.117181461)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 128.363 usec -> 2023-10-16T07:22:13.117.183.087 (XO: 9.117184186)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 125.763 usec -> 2023-10-16T07:22:13.117.180.487 (XO: 9.117181586)

 - Block number: 2 starting at 2023-10-16T07:22:23.856.464.725 (XO: 19.856468537 - 19.856774538, t=38.8 degC)  Rank: 00075
   TM values: Max =   212  Min =  -272  Ave =    -4.54  Std =    24.47
   Min TM value:-272 at an offset of 115.263 usec -> 2023-10-16T07:22:23.856.579.987 (XO: 19.856583800)
   Max TM value:212 at an offset of 114.963 usec -> 2023-10-16T07:22:23.856.579.687 (XO: 19.856583500)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 117.800 usec -> 2023-10-16T07:22:23.856.582.525 (XO: 19.856586337)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 103.063 usec -> 2023-10-16T07:22:23.856.567.787 (XO: 19.856571600)
