-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 22:29:58 2024 UTC
-- File 20231016_071742_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.438708 to 95.335046
-each separately from 1s to 95s and from 8137us to 807942us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.569140 in 1 packets: 18
      13.799227 in 1 packets: 36
      10.989253 in 1 packets: 54
      28.278385 in 1 packets: 72
      11.659141 in 1 packets: 90
      9.599169 in 1 packets: 108
MIU packet time from 2023-10-16T07:17:46.438.708.000 to 2023-10-16T07:19:20.335.046.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.569140 in 1 packets: 18
      13.799227 in 1 packets: 36
      10.989253 in 1 packets: 54
      28.278385 in 1 packets: 72
      11.659141 in 1 packets: 90
      9.599169 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7708954 to 96.7705745
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999317 in 1 packets: 18
      12.9999557 in 1 packets: 36
      10.9999624 in 2 packets: 54, 90
      28.9999010 in 1 packets: 72
      9.9999659 in 1 packets: 108
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C2 -> 38.8 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20231016_071742_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.2096048 to 3.2098938
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5690710  -- OK for SL --
   MIU time from 2023-10-16T07:17:46.438.708.000 to 2023-10-16T07:17:46.438.997.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.569140

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7789648 to 22.7792538
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7991830  -- OK for SL --
   MIU time from 2023-10-16T07:18:06.008.137.000 to 2023-10-16T07:18:06.008.426.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.799227

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.5784368 to 36.5787258
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9892150  -- OK for SL --
   MIU time from 2023-10-16T07:18:19.807.653.000 to 2023-10-16T07:18:19.807.942.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.989253

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.5679408 to 47.5682298
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.2782870  -- OK for SL --
   MIU time from 2023-10-16T07:18:30.797.195.000 to 2023-10-16T07:18:30.797.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.278385

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.8465168 to 75.8468058
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6591030  -- OK for SL --
   MIU time from 2023-10-16T07:18:59.075.869.000 to 2023-10-16T07:18:59.076.158.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.659141

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 87.5059088 to 87.5061978
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5991350  -- OK for SL --
   MIU time from 2023-10-16T07:19:10.735.299.000 to 2023-10-16T07:19:10.735.588.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.599169

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 97.1053328 to 97.1056218
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-16T07:19:20.334.757.000 to 2023-10-16T07:19:20.335.046.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.209604800 corrected by 40bit 10MHz counter of 2.770895400 obtained at 1Hz counter of 1. Result: 1.438710892
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-16T07:17:46.438.567.725
   Max 1st sample MIU time among all SL packets: 2023-10-16T07:19:20.334.905.725
   Min 1st sample XO time among all SL packets: 1.438570616
   Max 1st sample XO time among all SL packets: 95.334908197
   TM values from -275 to 259 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-16T07:17:46.438.567.725 (XO: 1.438570616 - 1.438876617, t=38.8 degC)  Rank: 00071
   TM values: Max =   254  Min =  -262  Ave =    -4.52  Std =    35.47
   Min TM value:-262 at an offset of 118.200 usec -> 2023-10-16T07:17:46.438.685.925 (XO: 1.438688817)
   Max TM value:254 at an offset of 117.913 usec -> 2023-10-16T07:17:46.438.685.637 (XO: 1.438688529)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 120.625 usec -> 2023-10-16T07:17:46.438.688.350 (XO: 1.438691242)
   Min dTM/dt [1/us]:-2.47e+03 at an offset of 118.025 usec -> 2023-10-16T07:17:46.438.685.750 (XO: 1.438688642)

 - Block number: 2 starting at 2023-10-16T07:18:06.007.996.725 (XO: 21.007997452 - 21.008303453, t=38.8 degC)  Rank: 00083
   TM values: Max =   202  Min =  -236  Ave =    -4.56  Std =    22.08
   Min TM value:-236 at an offset of 118.038 usec -> 2023-10-16T07:18:06.008.114.762 (XO: 21.008115490)
   Max TM value:202 at an offset of 117.738 usec -> 2023-10-16T07:18:06.008.114.462 (XO: 21.008115190)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 120.650 usec -> 2023-10-16T07:18:06.008.117.375 (XO: 21.008118103)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 117.913 usec -> 2023-10-16T07:18:06.008.114.637 (XO: 21.008115365)

 - Block number: 3 starting at 2023-10-16T07:18:19.807.512.725 (XO: 34.807516551 - 34.807822552, t=38.8 degC)  Rank: 00080
   TM values: Max =   206  Min =  -275  Ave =    -4.43  Std =    20.44
   Min TM value:-275 at an offset of 120.438 usec -> 2023-10-16T07:18:19.807.633.162 (XO: 34.807636989)
   Max TM value:206 at an offset of 120.188 usec -> 2023-10-16T07:18:19.807.632.912 (XO: 34.807636739)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 117.988 usec -> 2023-10-16T07:18:19.807.630.712 (XO: 34.807634539)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 120.288 usec -> 2023-10-16T07:18:19.807.633.012 (XO: 34.807636839)

 - Block number: 4 starting at 2023-10-16T07:18:30.797.054.725 (XO: 45.797058035 - 45.797364036, t=38.8 degC)  Rank: 00080
   TM values: Max =   194  Min =  -240  Ave =    -4.52  Std =    26.10
   Min TM value:-240 at an offset of 121.550 usec -> 2023-10-16T07:18:30.797.176.275 (XO: 45.797179585)
   Max TM value:194 at an offset of 121.188 usec -> 2023-10-16T07:18:30.797.175.912 (XO: 45.797179223)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 124.225 usec -> 2023-10-16T07:18:30.797.178.950 (XO: 45.797182260)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 109.225 usec -> 2023-10-16T07:18:30.797.163.950 (XO: 45.797167260)

 - Block number: 5 starting at 2023-10-16T07:18:59.075.728.725 (XO: 74.075730582 - 74.076036584, t=38.8 degC)  Rank: 00079
   TM values: Max =   206  Min =  -243  Ave =    -4.51  Std =    30.44
   Min TM value:-243 at an offset of 111.388 usec -> 2023-10-16T07:18:59.075.840.112 (XO: 74.075841970)
   Max TM value:206 at an offset of 108.725 usec -> 2023-10-16T07:18:59.075.837.450 (XO: 74.075839308)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 111.500 usec -> 2023-10-16T07:18:59.075.840.225 (XO: 74.075842083)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 138.938 usec -> 2023-10-16T07:18:59.075.867.662 (XO: 74.075869520)

 - Block number: 6 starting at 2023-10-16T07:19:10.735.158.725 (XO: 85.735162425 - 85.735468426, t=38.8 degC)  Rank: 00079
   TM values: Max =   205  Min =  -245  Ave =    -4.52  Std =    27.77
   Min TM value:-245 at an offset of 123.038 usec -> 2023-10-16T07:19:10.735.281.762 (XO: 85.735285462)
   Max TM value:205 at an offset of 151.126 usec -> 2023-10-16T07:19:10.735.309.850 (XO: 85.735313550)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 123.150 usec -> 2023-10-16T07:19:10.735.281.875 (XO: 85.735285575)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 151.238 usec -> 2023-10-16T07:19:10.735.309.963 (XO: 85.735313663)

 - Block number: 7 starting at 2023-10-16T07:19:20.334.616.725 (XO: 95.334619196 - 95.334925197, t=38.8 degC)  Rank: 00089
   TM values: Max =   259  Min =  -272  Ave =    -4.45  Std =    23.35
   Min TM value:-272 at an offset of 97.513 usec -> 2023-10-16T07:19:20.334.714.237 (XO: 95.334716709)
   Max TM value:259 at an offset of 93.200 usec -> 2023-10-16T07:19:20.334.709.925 (XO: 95.334712396)
   Max dTM/dt [1/us]:2.85e+03 at an offset of 93.038 usec -> 2023-10-16T07:19:20.334.709.762 (XO: 95.334712234)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 95.113 usec -> 2023-10-16T07:19:20.334.711.837 (XO: 95.334714309)
