-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 22:21:41 2024 UTC
-- File 20231016_070621_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.202235 to 21.132548
-each separately from 14s to 21s and from 132259us to 202524us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.929735 in 1 packets: 18
MIU packet time from 2023-10-16T07:06:38.202.235.000 to 2023-10-16T07:06:45.132.548.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.929735 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8343553 to 22.8343315
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999762 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231016_070621_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.0365914 to 16.0368804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9297110  -- OK for SL --
   MIU time from 2023-10-16T07:06:38.202.235.000 to 2023-10-16T07:06:38.202.524.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.929735

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9665914 to 22.9668804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-16T07:06:45.132.259.000 to 2023-10-16T07:06:45.132.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.036591400 corrected by 40bit 10MHz counter of 15.834355300 obtained at 1Hz counter of 14. Result: 14.202236767
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-16T07:06:38.202.094.725
   Max 1st sample MIU time among all SL packets: 2023-10-16T07:06:45.132.407.725
   Min 1st sample XO time among all SL packets: 14.202096492
   Max 1st sample XO time among all SL packets: 21.132409088
   TM values from -279 to 214 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-16T07:06:38.202.094.725 (XO: 14.202096492 - 14.202402493, t=38.8 degC)  Rank: 00081
   TM values: Max =   214  Min =  -279  Ave =    -4.47  Std =    23.47
   Min TM value:-279 at an offset of 122.138 usec -> 2023-10-16T07:06:38.202.216.862 (XO: 14.202218630)
   Max TM value:214 at an offset of 109.488 usec -> 2023-10-16T07:06:38.202.204.212 (XO: 14.202205980)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 124.738 usec -> 2023-10-16T07:06:38.202.219.462 (XO: 14.202221230)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 109.663 usec -> 2023-10-16T07:06:38.202.204.387 (XO: 14.202206155)

 - Block number: 2 starting at 2023-10-16T07:06:45.132.118.725 (XO: 21.132120087 - 21.132426088, t=38.8 degC)  Rank: 00077
   TM values: Max =   185  Min =  -255  Ave =    -4.43  Std =    22.21
   Min TM value:-255 at an offset of 101.500 usec -> 2023-10-16T07:06:45.132.220.225 (XO: 21.132221588)
   Max TM value:185 at an offset of 104.175 usec -> 2023-10-16T07:06:45.132.222.900 (XO: 21.132224263)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 104.000 usec -> 2023-10-16T07:06:45.132.222.725 (XO: 21.132224088)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 88.850 usec -> 2023-10-16T07:06:45.132.207.575 (XO: 21.132208938)
