-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 22:18:50 2024 UTC
-- File 20231016_070315_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841253us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU packet time from 2023-10-16T07:03:18.000.000.000 to 2023-10-16T07:03:32.177.027.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7426635 to 15.7426158
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999966 in 9 packets: 2, 4, 6, 9, 10, 11, 12, 14, 17
      0.9999965 in 3 packets: 3, 8, 15
      0.9999967 in 2 packets: 5, 16
   XO PPM: 
      -3.4 in 9 packets: 2, 4, 6, 9, 10, 11, 12, 14, 17
      -3.5 in 3 packets: 3, 8, 15
      -3.3 in 2 packets: 5, 16
Temperature FPGA: 
      03 -> 51.1 degC in 5 packets: 1, 6, 12, 15, 18
      02 -> 52.4 degC in 12 packets: 2, 3, 4, 7, 8, 9, 10, 11, 13, 14, 16, 17
      04 -> 49.8 degC in 1 packets: 5
Temperature XO: 
      C2 -> 38.8 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 13 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14
      2 in 4 packets: 15, 16, 17, 18
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 13 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14
      2 in 4 packets: 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20231016_070315_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7426636 to 15.9196436
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-16T07:03:18.000.000.000 to 2023-10-16T07:03:32.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833943 in 8 packets: 2, 4, 6, 8, 10, 12, 15, 16
      0.833944 in 6 packets: 3, 5, 9, 11, 14, 17
      0.833940 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.742663600 corrected by 40bit 10MHz counter of 1.742663500 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996700
   Min 1st sample MIU time among all SR packets: 2023-10-16T07:03:17.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-10-16T07:03:32.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028327
   TM values from -48 to 40 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-10-16T07:03:17.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    24  Min =   -32  Ave =    -4.50  Std =    11.86

 - Packet number: 2 starting at 2023-10-16T07:03:18.833.938.925 (XO: 0.833942860 - 0.833959860, t=38.8 degC)
   TM values: Max =    29  Min =   -36  Ave =    -4.46  Std =    13.91

 - Packet number: 3 starting at 2023-10-16T07:03:19.667.881.925 (XO: 1.667885763 - 1.667902763, t=38.8 degC)
   TM values: Max =    18  Min =   -28  Ave =    -4.52  Std =     9.55

 - Packet number: 4 starting at 2023-10-16T07:03:20.501.825.925 (XO: 2.501828631 - 2.501845631, t=38.8 degC)
   TM values: Max =    13  Min =   -23  Ave =    -4.55  Std =     5.96

 - Packet number: 5 starting at 2023-10-16T07:03:21.335.768.925 (XO: 3.335771433 - 3.335788433, t=38.8 degC)
   TM values: Max =    40  Min =   -45  Ave =    -4.52  Std =    20.57

 - Packet number: 6 starting at 2023-10-16T07:03:22.169.712.925 (XO: 4.169714202 - 4.169731202, t=38.8 degC)
   TM values: Max =    37  Min =   -48  Ave =    -4.59  Std =    20.69

 - Packet number: 7 starting at 2023-10-16T07:03:23.003.655.925 (XO: 5.003657038 - 5.003674038, t=38.8 degC)
   TM values: Max =    35  Min =   -40  Ave =    -4.31  Std =    17.73

 - Packet number: 8 starting at 2023-10-16T07:03:23.837.595.925 (XO: 5.837599957 - 5.837616957, t=38.8 degC)
   TM values: Max =    27  Min =   -39  Ave =    -4.69  Std =    13.45

 - Packet number: 9 starting at 2023-10-16T07:03:24.671.538.925 (XO: 6.671542808 - 6.671559808, t=38.8 degC)
   TM values: Max =    10  Min =   -20  Ave =    -4.50  Std =     5.35

 - Packet number: 10 starting at 2023-10-16T07:03:25.505.482.925 (XO: 7.505485644 - 7.505502644, t=38.8 degC)
   TM values: Max =    27  Min =   -37  Ave =    -4.49  Std =    15.89

 - Packet number: 11 starting at 2023-10-16T07:03:26.339.425.925 (XO: 8.339428479 - 8.339445479, t=38.8 degC)
   TM values: Max =    26  Min =   -38  Ave =    -4.50  Std =    13.93

 - Packet number: 12 starting at 2023-10-16T07:03:27.173.369.925 (XO: 9.173371314 - 9.173388315, t=38.8 degC)
   TM values: Max =    21  Min =   -31  Ave =    -4.54  Std =     9.19

 - Packet number: 13 starting at 2023-10-16T07:03:28.007.312.925 (XO: 10.007314150 - 10.007331150, t=38.8 degC)
   TM values: Max =    25  Min =   -32  Ave =    -4.41  Std =    10.81

 - Packet number: 14 starting at 2023-10-16T07:03:28.841.252.925 (XO: 10.841256985 - 10.841273985, t=38.8 degC)
   TM values: Max =    17  Min =   -24  Ave =    -4.50  Std =     7.04

 - Packet number: 15 starting at 2023-10-16T07:03:29.675.196.925 (XO: 11.675199888 - 11.675216888, t=38.8 degC)
   TM values: Max =    18  Min =   -28  Ave =    -4.54  Std =    10.22

 - Packet number: 16 starting at 2023-10-16T07:03:30.509.139.925 (XO: 12.509142705 - 12.509159705, t=38.8 degC)
   TM values: Max =    21  Min =   -33  Ave =    -4.60  Std =    11.61

 - Packet number: 17 starting at 2023-10-16T07:03:31.343.082.925 (XO: 13.343085491 - 13.343102492, t=38.8 degC)
   TM values: Max =    29  Min =   -37  Ave =    -4.53  Std =    15.23

 - Packet number: 18 starting at 2023-10-16T07:03:32.177.026.925 (XO: 14.177028327 - 14.177045327, t=38.8 degC)
   TM values: Max =    29  Min =   -40  Ave =    -4.52  Std =    16.60
