-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 21:22:06 2024 UTC
-- File 20231016_044840_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.895346 to 22.665429
-each separately from 7s to 22s and from 665140us to 895635us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.769505 in 1 packets: 18
MIU packet time from 2023-10-16T04:48:50.895.346.000 to 2023-10-16T04:49:05.665.429.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.769505 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7892227 to 23.7891720
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999493 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231016_044840_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6845693 to 9.6848583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7694550  -- OK for SL --
   MIU time from 2023-10-16T04:48:50.895.346.000 to 2023-10-16T04:48:50.895.635.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.769505

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4543133 to 24.4546023
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-16T04:49:05.665.140.000 to 2023-10-16T04:49:05.665.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.684569300 corrected by 40bit 10MHz counter of 8.789222700 obtained at 1Hz counter of 7. Result: 7.895349644
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-16T04:48:50.895.205.725
   Max 1st sample MIU time among all SL packets: 2023-10-16T04:49:05.665.288.725
   Min 1st sample XO time among all SL packets: 7.895209369
   Max 1st sample XO time among all SL packets: 22.665292287
   TM values from -248 to 227 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-16T04:48:50.895.205.725 (XO: 7.895209369 - 7.895515370, t=38.8 degC)  Rank: 00079
   TM values: Max =   197  Min =  -246  Ave =    -4.40  Std =    23.96
   Min TM value:-246 at an offset of 113.963 usec -> 2023-10-16T04:48:50.895.319.687 (XO: 7.895323332)
   Max TM value:197 at an offset of 113.613 usec -> 2023-10-16T04:48:50.895.319.337 (XO: 7.895322982)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 116.475 usec -> 2023-10-16T04:48:50.895.322.200 (XO: 7.895325844)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 113.813 usec -> 2023-10-16T04:48:50.895.319.537 (XO: 7.895323182)

 - Block number: 2 starting at 2023-10-16T04:49:05.664.999.725 (XO: 22.665003286 - 22.665309287, t=38.8 degC)  Rank: 00079
   TM values: Max =   227  Min =  -248  Ave =    -4.40  Std =    25.18
   Min TM value:-248 at an offset of 127.900 usec -> 2023-10-16T04:49:05.665.127.625 (XO: 22.665131186)
   Max TM value:227 at an offset of 123.500 usec -> 2023-10-16T04:49:05.665.123.225 (XO: 22.665126786)
   Max dTM/dt [1/us]:2.34e+03 at an offset of 128.050 usec -> 2023-10-16T04:49:05.665.127.775 (XO: 22.665131336)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 125.413 usec -> 2023-10-16T04:49:05.665.125.137 (XO: 22.665128699)
