-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 27 03:07:33 2024 UTC
-- File 20231013_180714_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.284683 to 28.964495
-each separately from 4s to 28s and from 284683us to 964495us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.679234 in 1 packets: 18
MIU packet time from 2023-10-13T18:07:21.284.683.000 to 2023-10-13T18:07:45.964.495.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.679234 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8265852 to 29.8265030
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.9999178 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231013_180714_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.1112697 to 6.1115587
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.6791510  -- OK for SL --
   MIU time from 2023-10-13T18:07:21.284.683.000 to 2023-10-13T18:07:21.284.972.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.679234

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.7907097 to 30.7909987
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-13T18:07:45.964.206.000 to 2023-10-13T18:07:45.964.495.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.111269700 corrected by 40bit 10MHz counter of 5.826585200 obtained at 1Hz counter of 4. Result: 4.284685496
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-13T18:07:21.284.542.725
   Max 1st sample MIU time among all SL packets: 2023-10-13T18:07:45.964.354.725
   Min 1st sample XO time among all SL packets: 4.284545221
   Max 1st sample XO time among all SL packets: 28.964358704
   TM values from -284 to 240 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-13T18:07:21.284.542.725 (XO: 4.284545221 - 4.284851222, t=38.8 degC)  Rank: 00080
   TM values: Max =   211  Min =  -280  Ave =    -4.66  Std =    26.08
   Min TM value:-280 at an offset of 125.450 usec -> 2023-10-13T18:07:21.284.668.175 (XO: 4.284670671)
   Max TM value:211 at an offset of 125.088 usec -> 2023-10-13T18:07:21.284.667.812 (XO: 4.284670309)
   Max dTM/dt [1/us]:2.13e+03 at an offset of 127.975 usec -> 2023-10-13T18:07:21.284.670.700 (XO: 4.284673196)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 125.225 usec -> 2023-10-13T18:07:21.284.667.950 (XO: 4.284670446)

 - Block number: 2 starting at 2023-10-13T18:07:45.964.065.725 (XO: 28.964069703 - 28.964375704, t=38.8 degC)  Rank: 00085
   TM values: Max =   240  Min =  -284  Ave =    -4.54  Std =    25.30
   Min TM value:-284 at an offset of 127.263 usec -> 2023-10-13T18:07:45.964.192.987 (XO: 28.964196966)
   Max TM value:240 at an offset of 126.988 usec -> 2023-10-13T18:07:45.964.192.712 (XO: 28.964196691)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 129.675 usec -> 2023-10-13T18:07:45.964.195.400 (XO: 28.964199378)
   Min dTM/dt [1/us]:-2.91e+03 at an offset of 127.100 usec -> 2023-10-13T18:07:45.964.192.825 (XO: 28.964196803)
