-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 26 22:42:26 2024 UTC
-- File 20231013_091559_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.835836 to 41.954433
-each separately from 6s to 41s and from 375060us to 954433us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.538935 in 1 packets: 18
      18.578795 in 1 packets: 36
MIU packet time from 2023-10-13T09:16:08.835.836.000 to 2023-10-13T09:16:43.954.433.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.538935 in 1 packets: 18
      18.578795 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9246461 to 42.9245292
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999432 in 1 packets: 18
      17.9999399 in 1 packets: 36
Temperature FPGA: 
      02 -> 52.4 degC in 54 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231013_091559_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7604830 to 8.7607720
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5388790  -- OK for SL --
   MIU time from 2023-10-13T09:16:08.835.836.000 to 2023-10-13T09:16:08.836.125.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.538935

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2996510 to 25.2999400
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5787350  -- OK for SL --
   MIU time from 2023-10-13T09:16:25.375.060.000 to 2023-10-13T09:16:25.375.349.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.578795

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.8786750 to 43.8789640
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-13T09:16:43.954.144.000 to 2023-10-13T09:16:43.954.433.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.760483000 corrected by 40bit 10MHz counter of 7.924646100 obtained at 1Hz counter of 6. Result: 6.835839742
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-13T09:16:08.835.695.725
   Max 1st sample MIU time among all SL packets: 2023-10-13T09:16:43.954.292.725
   Min 1st sample XO time among all SL packets: 6.835699466
   Max 1st sample XO time among all SL packets: 41.954297770
   TM values from -252 to 205 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-13T09:16:08.835.695.725 (XO: 6.835699466 - 6.836005467, t=40.4 degC)  Rank: 00083
   TM values: Max =   205  Min =  -252  Ave =    -4.58  Std =    18.50
   Min TM value:-252 at an offset of 130.713 usec -> 2023-10-13T09:16:08.835.826.437 (XO: 6.835830179)
   Max TM value:205 at an offset of 130.450 usec -> 2023-10-13T09:16:08.835.826.175 (XO: 6.835829917)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 133.125 usec -> 2023-10-13T09:16:08.835.828.850 (XO: 6.835832592)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 130.563 usec -> 2023-10-13T09:16:08.835.826.287 (XO: 6.835830029)

 - Block number: 2 starting at 2023-10-13T09:16:25.374.919.725 (XO: 23.374922700 - 23.375228701, t=40.4 degC)  Rank: 00080
   TM values: Max =   162  Min =  -225  Ave =    -4.52  Std =    17.52
   Min TM value:-225 at an offset of 92.025 usec -> 2023-10-13T09:16:25.375.011.750 (XO: 23.375014725)
   Max TM value:162 at an offset of 94.713 usec -> 2023-10-13T09:16:25.375.014.437 (XO: 23.375017413)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 94.563 usec -> 2023-10-13T09:16:25.375.014.287 (XO: 23.375017263)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 91.900 usec -> 2023-10-13T09:16:25.375.011.625 (XO: 23.375014600)

 - Block number: 3 starting at 2023-10-13T09:16:43.954.003.725 (XO: 41.954008769 - 41.954314770, t=40.4 degC)  Rank: 00082
   TM values: Max =   188  Min =  -225  Ave =    -4.61  Std =    27.61
   Min TM value:-225 at an offset of 127.625 usec -> 2023-10-13T09:16:43.954.131.350 (XO: 41.954136394)
   Max TM value:188 at an offset of 184.626 usec -> 2023-10-13T09:16:43.954.188.350 (XO: 41.954193394)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 130.100 usec -> 2023-10-13T09:16:43.954.133.825 (XO: 41.954138869)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 184.751 usec -> 2023-10-13T09:16:43.954.188.475 (XO: 41.954193519)
