-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 26 21:14:51 2024 UTC
-- File 20231013_072720_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.744555 to 91.081025
-each separately from 5s to 91s and from 63363us to 972450us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.046277 in 1 packets: 18
      28.180751 in 1 packets: 36
      8.214413 in 1 packets: 54
      14.569247 in 1 packets: 72
      19.306675 in 1 packets: 90
      4.017084 in 1 packets: 108
MIU packet time from 2023-10-13T07:27:28.744.555.000 to 2023-10-13T07:28:54.081.025.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.046277 in 1 packets: 18
      28.180751 in 1 packets: 36
      8.214413 in 1 packets: 54
      14.569247 in 1 packets: 72
      19.306675 in 1 packets: 90
      4.017084 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.9357622 to 92.9354693
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999625 in 1 packets: 18
      27.9999043 in 1 packets: 36
      8.9999693 in 1 packets: 54
      13.9999523 in 1 packets: 72
      19.9999323 in 1 packets: 90
      3.9999864 in 1 packets: 108
Temperature FPGA: 
      03 -> 51.1 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
      02 -> 52.4 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature XO: 
      C2 -> 38.8 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20231013_072720_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.6803186 to 7.6806076
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.0462390  -- OK for SL --
   MIU time from 2023-10-13T07:27:28.744.555.000 to 2023-10-13T07:27:28.744.844.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.046277

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7268466 to 18.7271356
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.1806550  -- OK for SL --
   MIU time from 2023-10-13T07:27:39.791.121.000 to 2023-10-13T07:27:39.791.410.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.180751

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.9077906 to 46.9080796
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2143830  -- OK for SL --
   MIU time from 2023-10-13T07:28:07.972.161.000 to 2023-10-13T07:28:07.972.450.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.214413

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.1224626 to 55.1227516
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5691990  -- OK for SL --
   MIU time from 2023-10-13T07:28:16.186.863.000 to 2023-10-13T07:28:16.187.152.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.569247

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.6919506 to 69.6922396
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3066070  -- OK for SL --
   MIU time from 2023-10-13T07:28:30.756.399.000 to 2023-10-13T07:28:30.756.688.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.306675

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 88.9988466 to 88.9991356
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0170710  -- OK for SL --
   MIU time from 2023-10-13T07:28:50.063.363.000 to 2023-10-13T07:28:50.063.652.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.017084

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 93.0162066 to 93.0164956
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-13T07:28:54.080.736.000 to 2023-10-13T07:28:54.081.025.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.680318600 corrected by 40bit 10MHz counter of 6.935762200 obtained at 1Hz counter of 5. Result: 5.744558932
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-13T07:27:28.744.414.725
   Max 1st sample MIU time among all SL packets: 2023-10-13T07:28:54.080.884.725
   Min 1st sample XO time among all SL packets: 5.744418656
   Max 1st sample XO time among all SL packets: 91.080886300
   TM values from -253 to 186 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-13T07:27:28.744.414.725 (XO: 5.744418656 - 5.744724657, t=38.8 degC)  Rank: 00076
   TM values: Max =   159  Min =  -191  Ave =    -4.50  Std =    20.68
   Min TM value:-191 at an offset of 126.913 usec -> 2023-10-13T07:27:28.744.541.637 (XO: 5.744545569)
   Max TM value:159 at an offset of 127.213 usec -> 2023-10-13T07:27:28.744.541.937 (XO: 5.744545869)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 127.075 usec -> 2023-10-13T07:27:28.744.541.800 (XO: 5.744545731)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 97.963 usec -> 2023-10-13T07:27:28.744.512.687 (XO: 5.744516619)

 - Block number: 2 starting at 2023-10-13T07:27:39.790.980.725 (XO: 16.790984393 - 16.791290395, t=38.8 degC)  Rank: 00070
   TM values: Max =   173  Min =  -214  Ave =    -4.59  Std =    23.57
   Min TM value:-214 at an offset of 124.988 usec -> 2023-10-13T07:27:39.791.105.712 (XO: 16.791109381)
   Max TM value:173 at an offset of 112.013 usec -> 2023-10-13T07:27:39.791.092.737 (XO: 16.791096406)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 127.475 usec -> 2023-10-13T07:27:39.791.108.200 (XO: 16.791111869)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 124.800 usec -> 2023-10-13T07:27:39.791.105.525 (XO: 16.791109194)

 - Block number: 3 starting at 2023-10-13T07:28:07.972.020.725 (XO: 44.972024630 - 44.972330631, t=38.8 degC)  Rank: 00077
   TM values: Max =   167  Min =  -234  Ave =    -4.76  Std =    19.04
   Min TM value:-234 at an offset of 111.713 usec -> 2023-10-13T07:28:07.972.132.437 (XO: 44.972136343)
   Max TM value:167 at an offset of 114.388 usec -> 2023-10-13T07:28:07.972.135.112 (XO: 44.972139018)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 114.250 usec -> 2023-10-13T07:28:07.972.134.975 (XO: 44.972138880)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 111.588 usec -> 2023-10-13T07:28:07.972.132.312 (XO: 44.972136218)

 - Block number: 4 starting at 2023-10-13T07:28:16.186.722.725 (XO: 53.186724660 - 53.187030661, t=38.8 degC)  Rank: 00080
   TM values: Max =   186  Min =  -253  Ave =    -4.66  Std =    23.46
   Min TM value:-253 at an offset of 124.050 usec -> 2023-10-13T07:28:16.186.846.775 (XO: 53.186848710)
   Max TM value:186 at an offset of 111.200 usec -> 2023-10-13T07:28:16.186.833.925 (XO: 53.186835860)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 126.525 usec -> 2023-10-13T07:28:16.186.849.250 (XO: 53.186851185)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 123.925 usec -> 2023-10-13T07:28:16.186.846.650 (XO: 53.186848585)

 - Block number: 5 starting at 2023-10-13T07:28:30.756.258.725 (XO: 67.756262296 - 67.756568297, t=38.8 degC)  Rank: 00083
   TM values: Max =   175  Min =  -248  Ave =    -4.55  Std =    19.34
   Min TM value:-248 at an offset of 113.513 usec -> 2023-10-13T07:28:30.756.372.237 (XO: 67.756375809)
   Max TM value:175 at an offset of 113.238 usec -> 2023-10-13T07:28:30.756.371.962 (XO: 67.756375534)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 115.975 usec -> 2023-10-13T07:28:30.756.374.700 (XO: 67.756378272)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 113.388 usec -> 2023-10-13T07:28:30.756.372.112 (XO: 67.756375684)

 - Block number: 6 starting at 2023-10-13T07:28:50.063.222.725 (XO: 87.063223640 - 87.063529641, t=38.8 degC)  Rank: 00092
   TM values: Max =   168  Min =  -242  Ave =    -4.59  Std =    20.03
   Min TM value:-242 at an offset of 124.563 usec -> 2023-10-13T07:28:50.063.347.287 (XO: 87.063348203)
   Max TM value:168 at an offset of 124.213 usec -> 2023-10-13T07:28:50.063.346.937 (XO: 87.063347853)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 127.113 usec -> 2023-10-13T07:28:50.063.349.837 (XO: 87.063350753)
   Min dTM/dt [1/us]:-1.81e+03 at an offset of 124.388 usec -> 2023-10-13T07:28:50.063.347.112 (XO: 87.063348028)

 - Block number: 7 starting at 2023-10-13T07:28:54.080.595.725 (XO: 91.080597299 - 91.080903300, t=38.8 degC)  Rank: 00079
   TM values: Max =   174  Min =  -237  Ave =    -4.50  Std =    18.39
   Min TM value:-237 at an offset of 125.100 usec -> 2023-10-13T07:28:54.080.720.825 (XO: 91.080722399)
   Max TM value:174 at an offset of 124.813 usec -> 2023-10-13T07:28:54.080.720.537 (XO: 91.080722112)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 127.588 usec -> 2023-10-13T07:28:54.080.723.312 (XO: 91.080724887)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 124.988 usec -> 2023-10-13T07:28:54.080.720.712 (XO: 91.080722287)
