-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 26 19:15:57 2024 UTC
-- File 20231013_040823_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.346838 to 25.587391
-each separately from 4s to 25s and from 346838us to 587391us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.239975 in 1 packets: 18
MIU packet time from 2023-10-13T04:08:30.346.838.000 to 2023-10-13T04:08:51.587.391.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.239975 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8993470 to 26.8992757
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999287 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231013_040823_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2461865 to 6.2464755
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2399030  -- OK for SL --
   MIU time from 2023-10-13T04:08:30.346.838.000 to 2023-10-13T04:08:30.347.127.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.239975

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4863785 to 27.4866675
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-13T04:08:51.587.102.000 to 2023-10-13T04:08:51.587.391.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.246186500 corrected by 40bit 10MHz counter of 5.899347000 obtained at 1Hz counter of 4. Result: 4.346840645
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-13T04:08:30.346.697.725
   Max 1st sample MIU time among all SL packets: 2023-10-13T04:08:51.587.250.725
   Min 1st sample XO time among all SL packets: 4.346700369
   Max 1st sample XO time among all SL packets: 25.587253522
   TM values from -250 to 186 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-13T04:08:30.346.697.725 (XO: 4.346700369 - 4.347006370, t=38.8 degC)  Rank: 00079
   TM values: Max =   186  Min =  -223  Ave =    -4.61  Std =    25.33
   Min TM value:-223 at an offset of 186.251 usec -> 2023-10-13T04:08:30.346.883.975 (XO: 4.346886620)
   Max TM value:186 at an offset of 185.963 usec -> 2023-10-13T04:08:30.346.883.688 (XO: 4.346886332)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 129.975 usec -> 2023-10-13T04:08:30.346.827.700 (XO: 4.346830345)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 186.113 usec -> 2023-10-13T04:08:30.346.883.838 (XO: 4.346886482)

 - Block number: 2 starting at 2023-10-13T04:08:51.586.961.725 (XO: 25.586964521 - 25.587270522, t=38.8 degC)  Rank: 00084
   TM values: Max =   180  Min =  -250  Ave =    -4.49  Std =    18.74
   Min TM value:-250 at an offset of 126.988 usec -> 2023-10-13T04:08:51.587.088.712 (XO: 25.587091509)
   Max TM value:180 at an offset of 126.700 usec -> 2023-10-13T04:08:51.587.088.425 (XO: 25.587091221)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 129.488 usec -> 2023-10-13T04:08:51.587.091.212 (XO: 25.587094009)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 126.850 usec -> 2023-10-13T04:08:51.587.088.575 (XO: 25.587091371)
