-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 26 16:21:18 2024 UTC
-- File 20231013_004143_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.682210 to 22.362460
-each separately from 5s to 22s and from 362171us to 682499us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.679672 in 1 packets: 18
MIU packet time from 2023-10-13T00:41:51.682.210.000 to 2023-10-13T00:42:08.362.460.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.679672 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8704009 to 23.8703436
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999427 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20231013_004143_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.5526119 to 7.5529009
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6796150  -- OK for SL --
   MIU time from 2023-10-13T00:41:51.682.210.000 to 2023-10-13T00:41:51.682.499.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.679672

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2325159 to 24.2328049
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-13T00:42:08.362.171.000 to 2023-10-13T00:42:08.362.460.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.552611900 corrected by 40bit 10MHz counter of 6.870400900 obtained at 1Hz counter of 5. Result: 5.682213251
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-13T00:41:51.682.069.725
   Max 1st sample MIU time among all SL packets: 2023-10-13T00:42:08.362.319.725
   Min 1st sample XO time among all SL packets: 5.682072976
   Max 1st sample XO time among all SL packets: 22.362322257
   TM values from -255 to 184 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-13T00:41:51.682.069.725 (XO: 5.682072976 - 5.682378977, t=38.8 degC)  Rank: 00082
   TM values: Max =   184  Min =  -255  Ave =    -4.46  Std =    22.77
   Min TM value:-255 at an offset of 127.638 usec -> 2023-10-13T00:41:51.682.197.362 (XO: 5.682200614)
   Max TM value:184 at an offset of 127.350 usec -> 2023-10-13T00:41:51.682.197.075 (XO: 5.682200326)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 130.088 usec -> 2023-10-13T00:41:51.682.199.812 (XO: 5.682203064)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 127.475 usec -> 2023-10-13T00:41:51.682.197.200 (XO: 5.682200451)

 - Block number: 2 starting at 2023-10-13T00:42:08.362.030.725 (XO: 22.362033256 - 22.362339257, t=38.8 degC)  Rank: 00081
   TM values: Max =   165  Min =  -214  Ave =    -4.43  Std =    19.24
   Min TM value:-214 at an offset of 128.025 usec -> 2023-10-13T00:42:08.362.158.750 (XO: 22.362161281)
   Max TM value:165 at an offset of 127.738 usec -> 2023-10-13T00:42:08.362.158.462 (XO: 22.362160994)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 130.563 usec -> 2023-10-13T00:42:08.362.161.287 (XO: 22.362163819)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 127.888 usec -> 2023-10-13T00:42:08.362.158.612 (XO: 22.362161144)
