-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 23 15:19:35 2024 UTC
-- File 20231008_015553_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.868284 to 33.194315
-each separately from 10s to 33s and from 194026us to 868573us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.858046 in 1 packets: 18
      13.467118 in 1 packets: 36
MIU packet time from 2023-10-08T01:56:06.868.284.000 to 2023-10-08T01:56:29.194.315.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.858046 in 1 packets: 18
      13.467118 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9475173 to 34.9474394
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999695 in 1 packets: 18
      13.9999526 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      04 -> 49.8 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C2 -> 38.8 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231008_015553_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.8158024 to 12.8160914
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.8580150  -- OK for SL --
   MIU time from 2023-10-08T01:56:06.868.284.000 to 2023-10-08T01:56:06.868.573.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.858046

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6741064 to 21.6743954
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4670710  -- OK for SL --
   MIU time from 2023-10-08T01:56:15.726.619.000 to 2023-10-08T01:56:15.726.908.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.467118

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      95 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.1414664 to 35.1417554
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-08T01:56:29.194.026.000 to 2023-10-08T01:56:29.194.315.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.815802400 corrected by 40bit 10MHz counter of 11.947517300 obtained at 1Hz counter of 10. Result: 10.868288052
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-10-08T01:56:06.868.143.725
   Max 1st sample MIU time among all SL packets: 2023-10-08T01:56:29.194.174.725
   Min 1st sample XO time among all SL packets: 10.868147777
   Max 1st sample XO time among all SL packets: 33.194176385
   TM values from -822 to 825 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-08T01:56:06.868.143.725 (XO: 10.868147777 - 10.868453778, t=38.8 degC)  Rank: 00076
   TM values: Max =   176  Min =  -221  Ave =    -4.62  Std =    18.32
   Min TM value:-221 at an offset of 127.338 usec -> 2023-10-08T01:56:06.868.271.062 (XO: 10.868275115)
   Max TM value:176 at an offset of 127.050 usec -> 2023-10-08T01:56:06.868.270.775 (XO: 10.868274827)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 124.875 usec -> 2023-10-08T01:56:06.868.268.600 (XO: 10.868272652)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 127.175 usec -> 2023-10-08T01:56:06.868.270.900 (XO: 10.868274952)

 - Block number: 2 starting at 2023-10-08T01:56:15.726.478.725 (XO: 19.726481795 - 19.726787796, t=38.8 degC)  Rank: 00078
   TM values: Max =   188  Min =  -229  Ave =    -4.58  Std =    19.47
   Min TM value:-229 at an offset of 124.813 usec -> 2023-10-08T01:56:15.726.603.537 (XO: 19.726606608)
   Max TM value:188 at an offset of 124.550 usec -> 2023-10-08T01:56:15.726.603.275 (XO: 19.726606345)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 127.275 usec -> 2023-10-08T01:56:15.726.606.000 (XO: 19.726609070)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 124.663 usec -> 2023-10-08T01:56:15.726.603.387 (XO: 19.726606458)

 - Block number: 3 starting at 2023-10-08T01:56:29.193.885.725 (XO: 33.193887384 - 33.194193385, t=38.8 degC)  Rank: 00095
   TM values: Max =   825  Min =  -822  Ave =    -4.57  Std =    32.73
   Min TM value:-822 at an offset of 105.200 usec -> 2023-10-08T01:56:29.193.990.925 (XO: 33.193992585)
   Max TM value:825 at an offset of 105.050 usec -> 2023-10-08T01:56:29.193.990.775 (XO: 33.193992435)
   Max dTM/dt [1/us]:9.77e+03 at an offset of 104.975 usec -> 2023-10-08T01:56:29.193.990.700 (XO: 33.193992360)
   Min dTM/dt [1/us]:-9.02e+03 at an offset of 105.088 usec -> 2023-10-08T01:56:29.193.990.812 (XO: 33.193992472)
