-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 20 20:09:08 2024 UTC
-- File 20231002_095159_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.230563 to 42.221757
-each separately from 6s to 42s and from 190824us to 230852us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.959972 in 1 packets: 18
      25.030355 in 1 packets: 36
MIU packet time from 2023-10-02T09:52:08.230.563.000 to 2023-10-02T09:52:44.221.757.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.959972 in 1 packets: 18
      25.030355 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.2394280 to 44.2393062
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999628 in 1 packets: 18
      24.9999154 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      02 -> 52.4 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20231002_095159_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.4699917 to 8.4702807
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9599350  -- OK for SL --
   MIU time from 2023-10-02T09:52:08.230.563.000 to 2023-10-02T09:52:08.230.852.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.959972

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4302157 to 19.4305047
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.0302710  -- OK for SL --
   MIU time from 2023-10-02T09:52:19.190.824.000 to 2023-10-02T09:52:19.191.113.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.030355

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.4607757 to 44.4610647
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-10-02T09:52:44.221.468.000 to 2023-10-02T09:52:44.221.757.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.469991700 corrected by 40bit 10MHz counter of 8.239428000 obtained at 1Hz counter of 6. Result: 6.230564484
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-10-02T09:52:08.230.422.725
   Max 1st sample MIU time among all SL packets: 2023-10-02T09:52:44.221.616.725
   Min 1st sample XO time among all SL packets: 6.230424208
   Max 1st sample XO time among all SL packets: 42.221618956
   TM values from -254 to 198 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-10-02T09:52:08.230.422.725 (XO: 6.230424208 - 6.230730209, t=40.4 degC)  Rank: 00079
   TM values: Max =   194  Min =  -217  Ave =    -4.51  Std =    25.03
   Min TM value:-217 at an offset of 124.213 usec -> 2023-10-02T09:52:08.230.546.937 (XO: 6.230548421)
   Max TM value:194 at an offset of 180.876 usec -> 2023-10-02T09:52:08.230.603.600 (XO: 6.230605084)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 126.675 usec -> 2023-10-02T09:52:08.230.549.400 (XO: 6.230550884)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 180.988 usec -> 2023-10-02T09:52:08.230.603.713 (XO: 6.230605197)

 - Block number: 2 starting at 2023-10-02T09:52:19.190.683.725 (XO: 17.190685273 - 17.190991274, t=40.4 degC)  Rank: 00078
   TM values: Max =   198  Min =  -234  Ave =    -4.56  Std =    33.35
   Min TM value:-234 at an offset of 270.051 usec -> 2023-10-02T09:52:19.190.953.775 (XO: 17.190955324)
   Max TM value:198 at an offset of 269.763 usec -> 2023-10-02T09:52:19.190.953.488 (XO: 17.190955037)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 125.713 usec -> 2023-10-02T09:52:19.190.809.437 (XO: 17.190810986)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 269.876 usec -> 2023-10-02T09:52:19.190.953.600 (XO: 17.190955149)

 - Block number: 3 starting at 2023-10-02T09:52:44.221.327.725 (XO: 42.221329955 - 42.221635956, t=40.4 degC)  Rank: 00082
   TM values: Max =   170  Min =  -254  Ave =    -4.64  Std =    22.45
   Min TM value:-254 at an offset of 122.513 usec -> 2023-10-02T09:52:44.221.450.237 (XO: 42.221452468)
   Max TM value:170 at an offset of 122.225 usec -> 2023-10-02T09:52:44.221.449.950 (XO: 42.221452181)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 124.975 usec -> 2023-10-02T09:52:44.221.452.700 (XO: 42.221454931)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 122.388 usec -> 2023-10-02T09:52:44.221.450.112 (XO: 42.221452343)
