-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 19 05:53:59 2024 UTC
-- File 20230916_225921_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.353103 to 16.763183
-each separately from 2s to 16s and from 353103us to 763183us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.409502 in 1 packets: 18
MIU packet time from 2023-09-16T22:59:26.353.103.000 to 2023-09-16T22:59:40.763.183.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.409502 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.7106144 to 17.7105667
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999523 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230916_225921_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0637180 to 4.0640070
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4094550  -- OK for SL --
   MIU time from 2023-09-16T22:59:26.353.103.000 to 2023-09-16T22:59:26.353.392.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.409502

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.4734620 to 18.4737510
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-16T22:59:40.762.894.000 to 2023-09-16T22:59:40.763.183.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.063718000 corrected by 40bit 10MHz counter of 3.710614400 obtained at 1Hz counter of 2. Result: 2.353104801
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-16T22:59:26.352.962.725
   Max 1st sample MIU time among all SL packets: 2023-09-16T22:59:40.763.042.725
   Min 1st sample XO time among all SL packets: 2.352964525
   Max 1st sample XO time among all SL packets: 16.763046619
   TM values from -233 to 227 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-16T22:59:26.352.962.725 (XO: 2.352964525 - 2.353270526, t=38.8 degC)  Rank: 00083
   TM values: Max =   213  Min =  -233  Ave =    -4.52  Std =    20.40
   Min TM value:-233 at an offset of 117.363 usec -> 2023-09-16T22:59:26.353.080.087 (XO: 2.353081888)
   Max TM value:213 at an offset of 105.175 usec -> 2023-09-16T22:59:26.353.067.900 (XO: 2.353069700)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 119.975 usec -> 2023-09-16T22:59:26.353.082.700 (XO: 2.353084500)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 117.175 usec -> 2023-09-16T22:59:26.353.079.900 (XO: 2.353081700)

 - Block number: 2 starting at 2023-09-16T22:59:40.762.753.725 (XO: 16.762757618 - 16.763063619, t=38.8 degC)  Rank: 00079
   TM values: Max =   227  Min =  -213  Ave =    -4.56  Std =    20.77
   Min TM value:-213 at an offset of 107.625 usec -> 2023-09-16T22:59:40.762.861.350 (XO: 16.762865244)
   Max TM value:227 at an offset of 107.375 usec -> 2023-09-16T22:59:40.762.861.100 (XO: 16.762864994)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 110.163 usec -> 2023-09-16T22:59:40.762.863.887 (XO: 16.762867781)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 107.513 usec -> 2023-09-16T22:59:40.762.861.237 (XO: 16.762865131)
