-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 17 02:54:21 2024 UTC
-- File 20230913_205735_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.277398 to 53.957597
-each separately from 14s to 53s and from 277398us to 957597us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.339643 in 1 packets: 18
      8.819742 in 1 packets: 36
      22.519658 in 1 packets: 54
MIU packet time from 2023-09-13T20:57:52.277.398.000 to 2023-09-13T20:58:31.957.597.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.339643 in 1 packets: 18
      8.819742 in 1 packets: 36
      22.519658 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 16.3046486 to 55.3045152
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999726 in 1 packets: 18
      8.9999693 in 1 packets: 36
      21.9999247 in 1 packets: 54
Temperature FPGA: 
      03 -> 51.1 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230913_205735_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5820481 to 16.5823371
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3396150  -- OK for SL --
   MIU time from 2023-09-13T20:57:52.277.398.000 to 2023-09-13T20:57:52.277.687.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.339643

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.9219521 to 24.9222411
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.8197110  -- OK for SL --
   MIU time from 2023-09-13T20:58:00.617.330.000 to 2023-09-13T20:58:00.617.619.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.819742

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.7419521 to 33.7422411
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5195830  -- OK for SL --
   MIU time from 2023-09-13T20:58:09.437.361.000 to 2023-09-13T20:58:09.437.650.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.519658

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.2618241 to 56.2621131
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-13T20:58:31.957.308.000 to 2023-09-13T20:58:31.957.597.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.582048100 corrected by 40bit 10MHz counter of 16.304648600 obtained at 1Hz counter of 14. Result: 14.277400471
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-13T20:57:52.277.257.725
   Max 1st sample MIU time among all SL packets: 2023-09-13T20:58:31.957.456.725
   Min 1st sample XO time among all SL packets: 14.277260195
   Max 1st sample XO time among all SL packets: 53.957460880
   TM values from -285 to 207 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-13T20:57:52.277.257.725 (XO: 14.277260195 - 14.277566196, t=38.8 degC)  Rank: 00076
   TM values: Max =   207  Min =  -264  Ave =    -4.62  Std =    25.30
   Min TM value:-264 at an offset of 114.213 usec -> 2023-09-13T20:57:52.277.371.937 (XO: 14.277374408)
   Max TM value:207 at an offset of 113.913 usec -> 2023-09-13T20:57:52.277.371.637 (XO: 14.277374108)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 116.713 usec -> 2023-09-13T20:57:52.277.374.437 (XO: 14.277376908)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 114.050 usec -> 2023-09-13T20:57:52.277.371.775 (XO: 14.277374246)

 - Block number: 2 starting at 2023-09-13T20:58:00.617.189.725 (XO: 22.617192723 - 22.617498724, t=38.8 degC)  Rank: 00078
   TM values: Max =   195  Min =  -285  Ave =    -4.52  Std =    28.20
   Min TM value:-285 at an offset of 199.451 usec -> 2023-09-13T20:58:00.617.389.175 (XO: 22.617392174)
   Max TM value:195 at an offset of 199.163 usec -> 2023-09-13T20:58:00.617.388.888 (XO: 22.617391887)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 201.951 usec -> 2023-09-13T20:58:00.617.391.675 (XO: 22.617394674)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 199.276 usec -> 2023-09-13T20:58:00.617.389.000 (XO: 22.617391999)

 - Block number: 3 starting at 2023-09-13T20:58:09.437.220.725 (XO: 31.437222855 - 31.437528856, t=38.8 degC)  Rank: 00074
   TM values: Max =   206  Min =  -250  Ave =    -4.51  Std =    30.36
   Min TM value:-250 at an offset of 101.013 usec -> 2023-09-13T20:58:09.437.321.737 (XO: 31.437323868)
   Max TM value:206 at an offset of 127.425 usec -> 2023-09-13T20:58:09.437.348.150 (XO: 31.437350281)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 130.188 usec -> 2023-09-13T20:58:09.437.350.912 (XO: 31.437353043)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 127.613 usec -> 2023-09-13T20:58:09.437.348.337 (XO: 31.437350468)

 - Block number: 4 starting at 2023-09-13T20:58:31.957.167.725 (XO: 53.957171879 - 53.957477880, t=38.8 degC)  Rank: 00081
   TM values: Max =   198  Min =  -236  Ave =    -4.51  Std =    24.07
   Min TM value:-236 at an offset of 127.863 usec -> 2023-09-13T20:58:31.957.295.587 (XO: 53.957299742)
   Max TM value:198 at an offset of 127.575 usec -> 2023-09-13T20:58:31.957.295.300 (XO: 53.957299455)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 130.338 usec -> 2023-09-13T20:58:31.957.298.062 (XO: 53.957302217)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 127.688 usec -> 2023-09-13T20:58:31.957.295.412 (XO: 53.957299567)
