-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 17 02:04:01 2024 UTC
-- File 20230913_183304_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.254388 to 21.280289
-each separately from 8s to 21s and from 254388us to 280289us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.025323 in 1 packets: 18
MIU packet time from 2023-09-13T18:33:15.254.388.000 to 2023-09-13T18:33:28.280.289.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.025323 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.5526563 to 22.5526119
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999556 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230913_183304_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8070450 to 9.8073340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0252790  -- OK for SL --
   MIU time from 2023-09-13T18:33:15.254.388.000 to 2023-09-13T18:33:15.254.677.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.025323

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      103 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.8326130 to 22.8329020
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-13T18:33:28.280.000.000 to 2023-09-13T18:33:28.280.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 9.807045000 corrected by 40bit 10MHz counter of 9.552656300 obtained at 1Hz counter of 8. Result: 8.254389565
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-13T18:33:15.254.247.725
   Max 1st sample MIU time among all SL packets: 2023-09-13T18:33:28.280.148.725
   Min 1st sample XO time among all SL packets: 8.254249289
   Max 1st sample XO time among all SL packets: 21.280150806
   TM values from -1173 to 350 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-13T18:33:15.254.247.725 (XO: 8.254249289 - 8.254555290, t=38.8 degC)  Rank: 00100
   TM values: Max =   350  Min =  -549  Ave =    -4.28  Std =    35.14
   Min TM value:-549 at an offset of 115.913 usec -> 2023-09-13T18:33:15.254.363.637 (XO: 8.254365202)
   Max TM value:350 at an offset of 117.100 usec -> 2023-09-13T18:33:15.254.364.825 (XO: 8.254366390)
   Max dTM/dt [1/us]:4.32e+03 at an offset of 116.150 usec -> 2023-09-13T18:33:15.254.363.875 (XO: 8.254365440)
   Min dTM/dt [1/us]:-3.67e+03 at an offset of 115.863 usec -> 2023-09-13T18:33:15.254.363.587 (XO: 8.254365152)

 - Block number: 2 starting at 2023-09-13T18:33:28.279.859.725 (XO: 21.279861805 - 21.280167806, t=38.8 degC)  Rank: 00103
   TM values: Max =   231  Min = -1173  Ave =    -4.53  Std =    36.42
   Min TM value:-1173 at an offset of 110.438 usec -> 2023-09-13T18:33:28.279.970.162 (XO: 21.279972242)
   Max TM value:231 at an offset of 111.088 usec -> 2023-09-13T18:33:28.279.970.812 (XO: 21.279972892)
   Max dTM/dt [1/us]:1.06e+04 at an offset of 110.500 usec -> 2023-09-13T18:33:28.279.970.225 (XO: 21.279972305)
   Min dTM/dt [1/us]:-8.01e+03 at an offset of 110.400 usec -> 2023-09-13T18:33:28.279.970.125 (XO: 21.279972205)
