-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 17 00:11:57 2024 UTC
-- File 20230913_141749_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.012729 to 22.845379
-each separately from 10s to 22s and from 12729us to 845379us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.832072 in 1 packets: 18
MIU packet time from 2023-09-13T14:18:02.012.729.000 to 2023-09-13T14:18:14.845.379.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.832072 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6983372 to 23.6982965
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999593 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230913_141749_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7110677 to 11.7113567
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8320310  -- OK for SL --
   MIU time from 2023-09-13T14:18:02.012.729.000 to 2023-09-13T14:18:02.013.018.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.832072

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5433877 to 24.5436767
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-13T14:18:14.845.090.000 to 2023-09-13T14:18:14.845.379.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 11.711067700 corrected by 40bit 10MHz counter of 11.698337200 obtained at 1Hz counter of 10. Result: 10.012730543
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-13T14:18:02.012.588.725
   Max 1st sample MIU time among all SL packets: 2023-09-13T14:18:14.845.238.725
   Min 1st sample XO time among all SL packets: 10.012590268
   Max 1st sample XO time among all SL packets: 22.845242799
   TM values from -243 to 213 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-13T14:18:02.012.588.725 (XO: 10.012590268 - 10.012896269, t=37.2 degC)  Rank: 00083
   TM values: Max =   213  Min =  -243  Ave =    -4.48  Std =    19.35
   Min TM value:-243 at an offset of 116.813 usec -> 2023-09-13T14:18:02.012.705.537 (XO: 10.012707081)
   Max TM value:213 at an offset of 103.788 usec -> 2023-09-13T14:18:02.012.692.512 (XO: 10.012694056)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 119.300 usec -> 2023-09-13T14:18:02.012.708.025 (XO: 10.012709568)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 103.900 usec -> 2023-09-13T14:18:02.012.692.625 (XO: 10.012694168)

 - Block number: 2 starting at 2023-09-13T14:18:14.844.949.725 (XO: 22.844953798 - 22.845259799, t=37.2 degC)  Rank: 00079
   TM values: Max =   179  Min =  -199  Ave =    -4.57  Std =    17.66
   Min TM value:-199 at an offset of 117.325 usec -> 2023-09-13T14:18:14.845.067.050 (XO: 22.845071123)
   Max TM value:179 at an offset of 101.800 usec -> 2023-09-13T14:18:14.845.051.525 (XO: 22.845055598)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 117.450 usec -> 2023-09-13T14:18:14.845.067.175 (XO: 22.845071248)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 101.913 usec -> 2023-09-13T14:18:14.845.051.637 (XO: 22.845055711)
