-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 16 23:51:01 2024 UTC
-- File 20230913_135956_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.025570 to 21.536519
-each separately from 1s to 21s and from 25570us to 536519us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.510371 in 1 packets: 18
MIU packet time from 2023-09-13T14:00:00.025.570.000 to 2023-09-13T14:00:20.536.519.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.510371 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.6915113 to 22.6914436
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999323 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230913_135956_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7170828 to 2.7173718
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.5103030  -- OK for SL --
   MIU time from 2023-09-13T14:00:00.025.570.000 to 2023-09-13T14:00:00.025.859.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.510371

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2276748 to 23.2279638
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-13T14:00:20.536.230.000 to 2023-09-13T14:00:20.536.519.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 2.717082800 corrected by 40bit 10MHz counter of 2.691511300 obtained at 1Hz counter of 1. Result: 1.025571587
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-13T14:00:00.025.429.725
   Max 1st sample MIU time among all SL packets: 2023-09-13T14:00:20.536.378.725
   Min 1st sample XO time among all SL packets: 1.025431311
   Max 1st sample XO time among all SL packets: 21.536381749
   TM values from -234 to 194 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-13T14:00:00.025.429.725 (XO: 1.025431311 - 1.025737313, t=37.2 degC)  Rank: 00080
   TM values: Max =   168  Min =  -234  Ave =    -4.52  Std =    18.25
   Min TM value:-234 at an offset of 117.200 usec -> 2023-09-13T14:00:00.025.546.925 (XO: 1.025548512)
   Max TM value:168 at an offset of 117.475 usec -> 2023-09-13T14:00:00.025.547.200 (XO: 1.025548787)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 117.313 usec -> 2023-09-13T14:00:00.025.547.037 (XO: 1.025548624)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 114.563 usec -> 2023-09-13T14:00:00.025.544.287 (XO: 1.025545874)

 - Block number: 2 starting at 2023-09-13T14:00:20.536.089.725 (XO: 21.536092748 - 21.536398749, t=37.2 degC)  Rank: 00081
   TM values: Max =   194  Min =  -223  Ave =    -4.56  Std =    20.23
   Min TM value:-223 at an offset of 73.700 usec -> 2023-09-13T14:00:20.536.163.425 (XO: 21.536166448)
   Max TM value:194 at an offset of 73.438 usec -> 2023-09-13T14:00:20.536.163.162 (XO: 21.536166185)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 103.125 usec -> 2023-09-13T14:00:20.536.192.850 (XO: 21.536195873)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 73.563 usec -> 2023-09-13T14:00:20.536.163.287 (XO: 21.536166310)
