-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 14 13:30:23 2024 UTC
-- File 20230910_111453_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.166264 to 36.286346
-each separately from 12s to 36s and from 166264us to 286346us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.039657 in 1 packets: 18
      21.079558 in 1 packets: 36
MIU packet time from 2023-09-10T11:15:08.166.264.000 to 2023-09-10T11:15:32.286.346.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.039657 in 1 packets: 18
      21.079558 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.6661941 to 37.6661131
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999898 in 1 packets: 18
      20.9999292 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230910_111453_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.8324589 to 13.8327479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.0396470  -- OK for SL --
   MIU time from 2023-09-10T11:15:08.166.264.000 to 2023-09-10T11:15:08.166.553.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.039657

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.8723949 to 16.8726839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0794870  -- OK for SL --
   MIU time from 2023-09-10T11:15:11.206.210.000 to 2023-09-10T11:15:11.206.499.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.079558

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9521709 to 37.9524599
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-10T11:15:32.286.057.000 to 2023-09-10T11:15:32.286.346.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.832458900 corrected by 40bit 10MHz counter of 13.666194100 obtained at 1Hz counter of 12. Result: 12.166265365
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-09-10T11:15:08.166.123.725
   Max 1st sample MIU time among all SL packets: 2023-09-10T11:15:32.286.205.725
   Min 1st sample XO time among all SL packets: 12.166125090
   Max 1st sample XO time among all SL packets: 36.286207498
   TM values from -226 to 220 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-10T11:15:08.166.123.725 (XO: 12.166125090 - 12.166431091, t=37.2 degC)  Rank: 00085
   TM values: Max =   193  Min =  -224  Ave =    -4.58  Std =    25.38
   Min TM value:-224 at an offset of 157.401 usec -> 2023-09-10T11:15:08.166.281.125 (XO: 12.166282490)
   Max TM value:193 at an offset of 128.775 usec -> 2023-09-10T11:15:08.166.252.500 (XO: 12.166253865)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 99.088 usec -> 2023-09-10T11:15:08.166.222.812 (XO: 12.166224178)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 128.900 usec -> 2023-09-10T11:15:08.166.252.625 (XO: 12.166253990)

 - Block number: 2 starting at 2023-09-10T11:15:11.206.069.725 (XO: 15.206071405 - 15.206377406, t=37.2 degC)  Rank: 00081
   TM values: Max =   220  Min =  -226  Ave =    -4.59  Std =    19.09
   Min TM value:-226 at an offset of 110.663 usec -> 2023-09-10T11:15:11.206.180.387 (XO: 15.206182068)
   Max TM value:220 at an offset of 96.238 usec -> 2023-09-10T11:15:11.206.165.962 (XO: 15.206167643)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 110.775 usec -> 2023-09-10T11:15:11.206.180.500 (XO: 15.206182180)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 96.338 usec -> 2023-09-10T11:15:11.206.166.062 (XO: 15.206167743)

 - Block number: 3 starting at 2023-09-10T11:15:32.285.916.725 (XO: 36.285918497 - 36.286224498, t=37.2 degC)  Rank: 00078
   TM values: Max =   177  Min =  -201  Ave =    -4.60  Std =    19.50
   Min TM value:-201 at an offset of 98.038 usec -> 2023-09-10T11:15:32.286.014.762 (XO: 36.286016535)
   Max TM value:177 at an offset of 82.163 usec -> 2023-09-10T11:15:32.285.998.887 (XO: 36.286000660)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 98.175 usec -> 2023-09-10T11:15:32.286.014.900 (XO: 36.286016672)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 82.275 usec -> 2023-09-10T11:15:32.285.999.000 (XO: 36.286000772)
