-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 13:25:30 2024 UTC
-- File 20230906_205242_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.247767 to 80.244076
-each separately from 0s to 80s and from 243787us to 810124us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.099189 in 1 packets: 18
      9.255966 in 1 packets: 36
      13.863792 in 1 packets: 54
      18.341965 in 1 packets: 72
      13.433663 in 1 packets: 90
MIU packet time from 2023-09-06T20:52:45.247.767.000 to 2023-09-06T20:54:05.244.076.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.099189 in 1 packets: 18
      9.255966 in 1 packets: 36
      13.863792 in 1 packets: 54
      18.341965 in 1 packets: 72
      13.433663 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7103444 to 81.7100686
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999139 in 1 packets: 18
      8.9999690 in 1 packets: 36
      13.9999517 in 1 packets: 54
      17.9999380 in 1 packets: 72
      13.9999516 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      03 -> 51.1 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C2 -> 38.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C1 -> 37.2 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 108 packets (all packets)
Number of generated alerts: 
      0 in 108 packets (all packets)
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_205242_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9581127 to 1.9584017
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.0991030  -- OK for SL --
   MIU time from 2023-09-06T20:52:45.247.767.000 to 2023-09-06T20:52:45.248.056.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.099189

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0575047 to 27.0577937
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2559350  -- OK for SL --
   MIU time from 2023-09-06T20:53:10.347.245.000 to 2023-09-06T20:53:10.347.534.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.255966

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.3137287 to 36.3140177
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8637430  -- OK for SL --
   MIU time from 2023-09-06T20:53:19.603.500.000 to 2023-09-06T20:53:19.603.789.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.863792

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.1777607 to 50.1780497
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3419030  -- OK for SL --
   MIU time from 2023-09-06T20:53:33.467.581.000 to 2023-09-06T20:53:33.467.870.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.341965

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.5199527 to 68.5202417
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4336150  -- OK for SL --
   MIU time from 2023-09-06T20:53:51.809.835.000 to 2023-09-06T20:53:51.810.124.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.433663

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 81.9538567 to 81.9541457
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T20:54:05.243.787.000 to 2023-09-06T20:54:05.244.076.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.958112700 corrected by 40bit 10MHz counter of 1.710344400 obtained at 1Hz counter of 0. Result: 0.247769142
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T20:52:45.247.626.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T20:54:05.243.935.725
   Min 1st sample XO time among all SL packets: 0.247628867
   Max 1st sample XO time among all SL packets: 80.243937654
   TM values from -267 to 232 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T20:52:45.247.626.725 (XO: 0.247628867 - 0.247934868, t=38.8 degC)  Rank: 00074
   TM values: Max =   205  Min =  -221  Ave =    -4.27  Std =    23.83
   Min TM value:-221 at an offset of 111.900 usec -> 2023-09-06T20:52:45.247.738.625 (XO: 0.247740767)
   Max TM value:205 at an offset of 97.438 usec -> 2023-09-06T20:52:45.247.724.162 (XO: 0.247726305)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 112.025 usec -> 2023-09-06T20:52:45.247.738.750 (XO: 0.247740892)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 109.363 usec -> 2023-09-06T20:52:45.247.736.087 (XO: 0.247738230)

 - Block number: 2 starting at 2023-09-06T20:53:10.347.104.725 (XO: 25.347107305 - 25.347413306, t=38.8 degC)  Rank: 00078
   TM values: Max =   206  Min =  -246  Ave =    -4.36  Std =    23.67
   Min TM value:-246 at an offset of 124.338 usec -> 2023-09-06T20:53:10.347.229.062 (XO: 25.347231643)
   Max TM value:206 at an offset of 124.588 usec -> 2023-09-06T20:53:10.347.229.312 (XO: 25.347231893)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 124.488 usec -> 2023-09-06T20:53:10.347.229.212 (XO: 25.347231793)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 121.713 usec -> 2023-09-06T20:53:10.347.226.437 (XO: 25.347229018)

 - Block number: 3 starting at 2023-09-06T20:53:19.603.359.725 (XO: 34.603363237 - 34.603669238, t=38.8 degC)  Rank: 00074
   TM values: Max =   232  Min =  -267  Ave =    -4.42  Std =    27.50
   Min TM value:-267 at an offset of 95.763 usec -> 2023-09-06T20:53:19.603.455.487 (XO: 34.603459000)
   Max TM value:232 at an offset of 66.750 usec -> 2023-09-06T20:53:19.603.426.475 (XO: 34.603429987)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 98.250 usec -> 2023-09-06T20:53:19.603.457.975 (XO: 34.603461487)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 95.563 usec -> 2023-09-06T20:53:19.603.455.287 (XO: 34.603458800)

 - Block number: 4 starting at 2023-09-06T20:53:33.467.440.725 (XO: 48.467443014 - 48.467749015, t=37.2 degC)  Rank: 00072
   TM values: Max =   221  Min =  -251  Ave =    -4.30  Std =    33.29
   Min TM value:-251 at an offset of 122.575 usec -> 2023-09-06T20:53:33.467.563.300 (XO: 48.467565590)
   Max TM value:221 at an offset of 122.838 usec -> 2023-09-06T20:53:33.467.563.562 (XO: 48.467565852)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 122.700 usec -> 2023-09-06T20:53:33.467.563.425 (XO: 48.467565715)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 120.025 usec -> 2023-09-06T20:53:33.467.560.750 (XO: 48.467563040)

 - Block number: 5 starting at 2023-09-06T20:53:51.809.694.725 (XO: 66.809698259 - 66.810004260, t=38.8 degC)  Rank: 00077
   TM values: Max =   221  Min =  -260  Ave =    -4.32  Std =    26.45
   Min TM value:-260 at an offset of 108.575 usec -> 2023-09-06T20:53:51.809.803.300 (XO: 66.809806834)
   Max TM value:221 at an offset of 108.838 usec -> 2023-09-06T20:53:51.809.803.562 (XO: 66.809807097)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 108.688 usec -> 2023-09-06T20:53:51.809.803.412 (XO: 66.809806947)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 93.438 usec -> 2023-09-06T20:53:51.809.788.162 (XO: 66.809791697)

 - Block number: 6 starting at 2023-09-06T20:54:05.243.646.725 (XO: 80.243648653 - 80.243954654, t=37.2 degC)  Rank: 00072
   TM values: Max =   207  Min =  -255  Ave =    -4.33  Std =    24.89
   Min TM value:-255 at an offset of 106.463 usec -> 2023-09-06T20:54:05.243.753.187 (XO: 80.243755116)
   Max TM value:207 at an offset of 106.238 usec -> 2023-09-06T20:54:05.243.752.962 (XO: 80.243754891)
   Max dTM/dt [1/us]:2.00e+03 at an offset of 109.013 usec -> 2023-09-06T20:54:05.243.755.737 (XO: 80.243757666)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 106.325 usec -> 2023-09-06T20:54:05.243.753.050 (XO: 80.243754979)
