-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 13:20:31 2024 UTC
-- File 20230906_204612_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.433953 to 15.883676
-each separately from 12s to 15s and from 433953us to 883676us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.449145 in 1 packets: 18
MIU packet time from 2023-09-06T20:46:27.433.953.000 to 2023-09-06T20:46:30.883.676.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.449145 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7174030 to 16.7173927
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999897 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_204612_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.1513569 to 14.1516459
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.4491350  -- OK for SL --
   MIU time from 2023-09-06T20:46:27.433.953.000 to 2023-09-06T20:46:27.434.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.449145

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6007809 to 17.6010699
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T20:46:30.883.387.000 to 2023-09-06T20:46:30.883.676.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.151356900 corrected by 40bit 10MHz counter of 13.717403000 obtained at 1Hz counter of 12. Result: 12.433955375
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T20:46:27.433.812.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T20:46:30.883.535.725
   Min 1st sample XO time among all SL packets: 12.433815100
   Max 1st sample XO time among all SL packets: 15.883540017
   TM values from -272 to 220 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T20:46:27.433.812.725 (XO: 12.433815100 - 12.434121101, t=38.8 degC)  Rank: 00083
   TM values: Max =   220  Min =  -272  Ave =    -4.30  Std =    23.95
   Min TM value:-272 at an offset of 85.825 usec -> 2023-09-06T20:46:27.433.898.550 (XO: 12.433900925)
   Max TM value:220 at an offset of 100.863 usec -> 2023-09-06T20:46:27.433.913.587 (XO: 12.433915963)
   Max dTM/dt [1/us]:2.06e+03 at an offset of 100.750 usec -> 2023-09-06T20:46:27.433.913.475 (XO: 12.433915850)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 98.025 usec -> 2023-09-06T20:46:27.433.910.750 (XO: 12.433913125)

 - Block number: 2 starting at 2023-09-06T20:46:30.883.246.725 (XO: 15.883251016 - 15.883557017, t=38.8 degC)  Rank: 00075
   TM values: Max =   208  Min =  -222  Ave =    -4.32  Std =    36.30
   Min TM value:-222 at an offset of 179.251 usec -> 2023-09-06T20:46:30.883.425.975 (XO: 15.883430267)
   Max TM value:208 at an offset of 181.588 usec -> 2023-09-06T20:46:30.883.428.313 (XO: 15.883432605)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 179.338 usec -> 2023-09-06T20:46:30.883.426.063 (XO: 15.883430355)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 196.176 usec -> 2023-09-06T20:46:30.883.442.900 (XO: 15.883447192)
