-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 13:01:12 2024 UTC
-- File 20230906_201354_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177027
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 5 packets: 2, 5, 8, 10, 16
      0.833943 in 9 packets: 3, 4, 6, 9, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
MIU packet time from 2023-09-06T20:13:57.000.000.000 to 2023-09-06T20:14:11.177.027.000
   Differences in neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 5 packets: 2, 5, 8, 10, 16
      0.833943 in 9 packets: 3, 4, 6, 9, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.2840124 to 16.2839643
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999966 in 9 packets: 2, 3, 5, 8, 9, 12, 14, 16, 17
      0.9999965 in 5 packets: 4, 6, 10, 11, 15
   XO PPM: 
      -3.4 in 9 packets: 2, 3, 5, 8, 9, 12, 14, 16, 17
      -3.5 in 5 packets: 4, 6, 10, 11, 15
Temperature FPGA: 
      04 -> 49.8 degC in 12 packets: 1, 2, 5, 6, 9, 10, 12, 13, 14, 15, 16, 18
      05 -> 48.5 degC in 1 packets: 3
      03 -> 51.1 degC in 5 packets: 4, 7, 8, 11, 17
Temperature XO: 
      C2 -> 38.8 degC in 17 packets: 1, 2, 3, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C1 -> 37.2 degC in 1 packets: 4
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 2 packets: 17, 18
Number of generated alerts: 
      0 in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16
      1 in 2 packets: 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_201354_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2840125 to 16.4609925
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T20:13:57.000.000.000 to 2023-09-06T20:14:11.177.027.000
      Difference between neigboring packets: 
      0.833939 in 1 packets: 1
      0.833944 in 5 packets: 2, 5, 8, 10, 16
      0.833943 in 9 packets: 3, 4, 6, 9, 11, 12, 14, 15, 17
      0.833940 in 1 packets: 7
      0.833941 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.284012500 corrected by 40bit 10MHz counter of 2.284012400 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SR packets: 2023-09-06T20:13:56.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-09-06T20:14:11.177.026.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177028727
   TM values from -71 to 72 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-09-06T20:13:56.999.999.925 (XO: 0.000000025 - 0.000017025, t=38.8 degC)
   TM values: Max =    40  Min =   -45  Ave =    -4.18  Std =    14.46

 - Packet number: 2 starting at 2023-09-06T20:13:57.833.938.925 (XO: 0.833942860 - 0.833959860, t=38.8 degC)
   TM values: Max =    49  Min =   -58  Ave =    -4.44  Std =    17.41

 - Packet number: 3 starting at 2023-09-06T20:13:58.667.882.925 (XO: 1.667885696 - 1.667902696, t=38.8 degC)
   TM values: Max =    40  Min =   -55  Ave =    -4.40  Std =    16.02

 - Packet number: 4 starting at 2023-09-06T20:13:59.501.825.925 (XO: 2.501828581 - 2.501845581, t=37.2 degC)
   TM values: Max =    56  Min =   -67  Ave =    -4.38  Std =    22.74

 - Packet number: 5 starting at 2023-09-06T20:14:00.335.768.925 (XO: 3.335771467 - 3.335788467, t=38.8 degC)
   TM values: Max =    54  Min =   -53  Ave =    -4.67  Std =    17.93

 - Packet number: 6 starting at 2023-09-06T20:14:01.169.712.925 (XO: 4.169714319 - 4.169731319, t=38.8 degC)
   TM values: Max =    36  Min =   -46  Ave =    -4.09  Std =    14.47

 - Packet number: 7 starting at 2023-09-06T20:14:02.003.655.925 (XO: 5.003657237 - 5.003674237, t=38.8 degC)
   TM values: Max =    72  Min =   -70  Ave =    -4.62  Std =    27.26

 - Packet number: 8 starting at 2023-09-06T20:14:02.837.595.925 (XO: 5.837600073 - 5.837617073, t=38.8 degC)
   TM values: Max =    58  Min =   -71  Ave =    -4.45  Std =    24.47

 - Packet number: 9 starting at 2023-09-06T20:14:03.671.539.925 (XO: 6.671542908 - 6.671559908, t=38.8 degC)
   TM values: Max =    39  Min =   -49  Ave =    -4.38  Std =    15.92

 - Packet number: 10 starting at 2023-09-06T20:14:04.505.482.925 (XO: 7.505485794 - 7.505502794, t=38.8 degC)
   TM values: Max =    45  Min =   -56  Ave =    -4.04  Std =    19.20

 - Packet number: 11 starting at 2023-09-06T20:14:05.339.426.925 (XO: 8.339428713 - 8.339445713, t=38.8 degC)
   TM values: Max =    40  Min =   -45  Ave =    -4.68  Std =    15.87

 - Packet number: 12 starting at 2023-09-06T20:14:06.173.369.925 (XO: 9.173371614 - 9.173388615, t=38.8 degC)
   TM values: Max =    41  Min =   -52  Ave =    -4.40  Std =    15.87

 - Packet number: 13 starting at 2023-09-06T20:14:07.007.312.925 (XO: 10.007314450 - 10.007331450, t=38.8 degC)
   TM values: Max =    51  Min =   -54  Ave =    -4.77  Std =    18.59

 - Packet number: 14 starting at 2023-09-06T20:14:07.841.253.925 (XO: 10.841257285 - 10.841274285, t=38.8 degC)
   TM values: Max =    50  Min =   -60  Ave =    -4.48  Std =    19.11

 - Packet number: 15 starting at 2023-09-06T20:14:08.675.196.925 (XO: 11.675200188 - 11.675217188, t=38.8 degC)
   TM values: Max =    58  Min =   -59  Ave =    -4.27  Std =    21.84

 - Packet number: 16 starting at 2023-09-06T20:14:09.509.139.925 (XO: 12.509143056 - 12.509160056, t=38.8 degC)
   TM values: Max =    57  Min =   -71  Ave =    -4.59  Std =    23.09

 - Packet number: 17 starting at 2023-09-06T20:14:10.343.083.925 (XO: 13.343085891 - 13.343102892, t=38.8 degC)
   TM values: Max =    41  Min =   -45  Ave =    -4.23  Std =    15.16

 - Packet number: 18 starting at 2023-09-06T20:14:11.177.026.925 (XO: 14.177028727 - 14.177045727, t=38.8 degC)
   TM values: Max =    52  Min =   -47  Ave =    -4.47  Std =    16.00
