-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 12:06:57 2024 UTC
-- File 20230906_174820_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.955570 to 55.593061
-each separately from 8s to 55s and from 405571us to 955859us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.826721 in 1 packets: 18
      8.622702 in 1 packets: 36
      19.186912 in 1 packets: 54
MIU packet time from 2023-09-06T17:48:31.955.570.000 to 2023-09-06T17:49:18.593.061.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.826721 in 1 packets: 18
      8.622702 in 1 packets: 36
      19.186912 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.7221588 to 56.7219959
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999342 in 2 packets: 18, 54
      8.9999687 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_174820_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6777293 to 10.6780183
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8266550  -- OK for SL --
   MIU time from 2023-09-06T17:48:31.955.570.000 to 2023-09-06T17:48:31.955.859.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.826721

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.5046733 to 29.5049623
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6226710  -- OK for SL --
   MIU time from 2023-09-06T17:48:50.782.580.000 to 2023-09-06T17:48:50.782.869.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.622702

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.1276333 to 38.1279223
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.1868470  -- OK for SL --
   MIU time from 2023-09-06T17:48:59.405.571.000 to 2023-09-06T17:48:59.405.860.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.186912

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.3147693 to 57.3150583
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T17:49:18.592.772.000 to 2023-09-06T17:49:18.593.061.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.677729300 corrected by 40bit 10MHz counter of 9.722158800 obtained at 1Hz counter of 8. Result: 8.955573845
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-09-06T17:48:31.955.429.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T17:49:18.592.920.725
   Min 1st sample XO time among all SL packets: 8.955433569
   Max 1st sample XO time among all SL packets: 55.592924141
   TM values from -295 to 278 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T17:48:31.955.429.725 (XO: 8.955433569 - 8.955739570, t=37.2 degC)  Rank: 00071
   TM values: Max =   278  Min =  -295  Ave =    -4.41  Std =    31.28
   Min TM value:-295 at an offset of 102.938 usec -> 2023-09-06T17:48:31.955.532.662 (XO: 8.955536507)
   Max TM value:278 at an offset of 102.650 usec -> 2023-09-06T17:48:31.955.532.375 (XO: 8.955536219)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 105.475 usec -> 2023-09-06T17:48:31.955.535.200 (XO: 8.955539044)
   Min dTM/dt [1/us]:-2.46e+03 at an offset of 102.813 usec -> 2023-09-06T17:48:31.955.532.537 (XO: 8.955536382)

 - Block number: 2 starting at 2023-09-06T17:48:50.782.439.725 (XO: 27.782442764 - 27.782748765, t=37.2 degC)  Rank: 00071
   TM values: Max =   232  Min =  -255  Ave =    -4.42  Std =    25.63
   Min TM value:-255 at an offset of 112.150 usec -> 2023-09-06T17:48:50.782.551.875 (XO: 27.782554914)
   Max TM value:232 at an offset of 112.388 usec -> 2023-09-06T17:48:50.782.552.112 (XO: 27.782555151)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 112.275 usec -> 2023-09-06T17:48:50.782.552.000 (XO: 27.782555039)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 109.613 usec -> 2023-09-06T17:48:50.782.549.337 (XO: 27.782552376)

 - Block number: 3 starting at 2023-09-06T17:48:59.405.430.725 (XO: 36.405432703 - 36.405738705, t=37.2 degC)  Rank: 00079
   TM values: Max =   215  Min =  -259  Ave =    -4.30  Std =    29.08
   Min TM value:-259 at an offset of 115.188 usec -> 2023-09-06T17:48:59.405.545.912 (XO: 36.405547891)
   Max TM value:215 at an offset of 115.425 usec -> 2023-09-06T17:48:59.405.546.150 (XO: 36.405548129)
   Max dTM/dt [1/us]:2.51e+03 at an offset of 115.300 usec -> 2023-09-06T17:48:59.405.546.025 (XO: 36.405548004)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 112.600 usec -> 2023-09-06T17:48:59.405.543.325 (XO: 36.405545304)

 - Block number: 4 starting at 2023-09-06T17:49:18.592.631.725 (XO: 55.592635140 - 55.592941141, t=37.2 degC)  Rank: 00083
   TM values: Max =   212  Min =  -264  Ave =    -4.36  Std =    21.62
   Min TM value:-264 at an offset of 125.388 usec -> 2023-09-06T17:49:18.592.757.112 (XO: 55.592760528)
   Max TM value:212 at an offset of 110.300 usec -> 2023-09-06T17:49:18.592.742.025 (XO: 55.592745440)
   Max dTM/dt [1/us]:2.44e+03 at an offset of 125.513 usec -> 2023-09-06T17:49:18.592.757.237 (XO: 55.592760653)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 110.413 usec -> 2023-09-06T17:49:18.592.742.137 (XO: 55.592745553)
