-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 12 11:52:20 2024 UTC
-- File 20230906_172923_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.165538 to 21.133891
-each separately from 12s to 21s and from 133602us to 165827us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.967775 in 1 packets: 18
MIU packet time from 2023-09-06T17:29:38.165.538.000 to 2023-09-06T17:29:47.133.891.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.967775 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.1158973 to 23.1158661
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999688 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230906_172923_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.2814367 to 14.2817257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9677430  -- OK for SL --
   MIU time from 2023-09-06T17:29:38.165.538.000 to 2023-09-06T17:29:38.165.827.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.967775

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2494687 to 23.2497577
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-09-06T17:29:47.133.602.000 to 2023-09-06T17:29:47.133.891.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.281436700 corrected by 40bit 10MHz counter of 14.115897300 obtained at 1Hz counter of 12. Result: 12.165539979
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-09-06T17:29:38.165.397.725
   Max 1st sample MIU time among all SL packets: 2023-09-06T17:29:47.133.750.725
   Min 1st sample XO time among all SL packets: 12.165399704
   Max 1st sample XO time among all SL packets: 21.133751793
   TM values from -263 to 217 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-09-06T17:29:38.165.397.725 (XO: 12.165399704 - 12.165705705, t=37.2 degC)  Rank: 00075
   TM values: Max =   194  Min =  -263  Ave =    -4.49  Std =    21.41
   Min TM value:-263 at an offset of 112.563 usec -> 2023-09-06T17:29:38.165.510.287 (XO: 12.165512267)
   Max TM value:194 at an offset of 112.800 usec -> 2023-09-06T17:29:38.165.510.525 (XO: 12.165512504)
   Max dTM/dt [1/us]:2.43e+03 at an offset of 112.688 usec -> 2023-09-06T17:29:38.165.510.412 (XO: 12.165512392)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 115.475 usec -> 2023-09-06T17:29:38.165.513.200 (XO: 12.165515179)

 - Block number: 2 starting at 2023-09-06T17:29:47.133.461.725 (XO: 21.133462792 - 21.133768793, t=37.2 degC)  Rank: 00090
   TM values: Max =   217  Min =  -257  Ave =    -4.44  Std =    25.19
   Min TM value:-257 at an offset of 124.425 usec -> 2023-09-06T17:29:47.133.586.150 (XO: 21.133587218)
   Max TM value:217 at an offset of 124.700 usec -> 2023-09-06T17:29:47.133.586.425 (XO: 21.133587493)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 124.588 usec -> 2023-09-06T17:29:47.133.586.312 (XO: 21.133587380)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 121.813 usec -> 2023-09-06T17:29:47.133.583.537 (XO: 21.133584605)
